Acidity Test of Transformer Insulating Oil
Discharging a Capacitor
Charging a CapacitorClosely Related Articles
Measurement of Resistance
Wheatstone Bridge Circuit Theory and Principle
Kelvin Bridge Circuit | Kelvin Double Bridge
Maxwell Bridge Inductance Capacitance Bridge
Anderson′s Bridge | Advantages Disadvantages of Anderson′s Bridge
Hay′s Bridge Circuit Theory Phasor Diagram Advantages Applications
Owens Bridge Circuit and Advantages
Schering Bridge Measurement of Capacitance using Schering Bridge
De Sauty Bridge
Heaviside Bridge Circuit
Blavier Test | Murray Loop Test | Varley Loop Test | Fisher Loop Test
Sensor | Types of Sensor
Temperature Sensor Temperature Measurement | Types of Temperature Sensor
Digital Frequency Meter
Electrical Measuring Instruments | Types Accuracy Precision Resolution Speed
Cathode Ray Oscilloscope | CRO
Errors in Measurement | Classification of Errors
Lissajous Patterns of CRO or Cathode Ray Oscilloscope
Frequency Limitation of an Oscilloscope
Permanent Magnet Moving Coil Instrument or PMMC Instrument
Characteristics of Sensors
Moving Iron Instrument
Induction Type Meters
Measurement of Voltage Current and Frequency by Oscilloscope
Electrostatic Type Instruments Construction Principle Torque Equation
What is Low Power Factor Wattmeter ?
Resistance Temperature Detector or RTD | Construction and Working Principle
Energy Meter with Lag Adjustment Devices
Rectifier Type Instrument | Construction Principle of Operation
Digital Storage Oscilloscope
Electrodynamometer Type Wattmeter
Thermocouple type Instruments Construction Principle of Operation
Double Beam Oscilloscope
Protection of Meters
Measurement of Three Phase Power
Bimetallic Strip Thermometer
Varmeter | Single Phase and Polyphase Varmeter
Thermocouple Temperature Measurement
Megger | Working Principle Types History Uses of Megger
Radiation Pyrometer | Types Working Principle
Weston Type Frequency Meter
Power Factor Meters | Electrodynamometer Type Power Factor Meter
Phase Sequence Indicator
Ammeter Working Principle and Types of Ammeter
Insulation Resistance Test and Polarization Index Test
Tan Delta Test |Loss Angle Test | Dissipation Factor Test
Electronic DC Voltmeter
Working Principle of Voltmeter and Types of Voltmeter
RC Phase Shift Oscillator
Digital Voltmeters Working Principle of Digital Voltmeter
Wien Bridge Oscillator
Measurement of Electrical Energy
Energy Meter Testing
Construction of AC Energy Meter
Watt Hour Meter
Vector Impedance Meter
How to Use a Digital Multimeter?
Voltage Controlled Oscillator
Measurement of Resistance
Measurement of Low Resistance (<1Ω)The major problem in measurement of low resistance values is the contact resistance or lead resistance of the measuring instruments, though being small in value is comparable to the resistance being measured and hence causes serious error.
Thus to eliminate this issue small valued resistance are constructed with four terminals. Two terminals are current terminals and other two are potential terminals. Figure below shows the construction of low resistance. The current is flown through current terminals C1 and C2 while the potential drop is measured across potential terminals V1 and V2. Hence we can find out the value of resistance under experiment in terms of V and I as indicated in the above figure. This method helps us to exclude the contact resistance due to current terminals and though contact resistance of potential terminals still comes into picture, it is very small fraction of high resistance potential circuit and hence induces negligible error. The methods employed for measurement of low resistances are:-
- Kelvin’s Double Bridge Method
- Potentiometer Method
- Ducter Ohmmeter.
Kelvin’s Double BridgeKelvin’s double bridge is a modification of simple Wheatstone bridge. Figure below shows the circuit diagram of Kelvin’s double bridge. As we can see in the above figure there are two sets of arms, one with resistances P and Q and other with resistances p and q. R is the unknown low resistance and S is a standard resistance. Here r represents the contact resistance between the unknown resistance and the standard resistance, whose effect we need to eliminate. For measurement we make the ratio P/Q equal to p/q and hence a balanced Wheatstone bridge is formed leading to null deflection in the galvanometer. Hence for a balanced bridge we can write Putting eqn 2 in 1 and solving and using P/Q = p/q, we get- Hence we see that by using balanced double arms we can eliminate the contact resistance completely and hence error due to it. To eliminate another error caused due to thermo-electric emf, we take another reading with battery connection reversed and finally take average of the two readings. This bridge is useful for resistances in range of 0.1µΩ to 1.0 Ω.
Ducter OhmmeterIt is an electromechanical instrument used for measurement of low resistances. It comprises of a permanent magnet similar to that of a PMMC instrument and two coils in between the magnetic field created by the poles of the magnet. The two coils are at right angles to each other and are free to rotate about the common axis. Figure below shows a Ducter Ohmmeter and the connections required to measure an unknown resistance R. One of the coil called current coil, is connected to current terminals C1 and C2, while the other coil called, voltage coil is connected to potential terminals V1 and V2. Voltage coil carries current proportional of the voltage drop across R and so is its torque produced. Current coil carries current proportional to the current flowing through R and so is its torque too. Both the torque acts in opposite direction and the indicator come to halt when the two are equal. This instrument is useful for resistance in range 100µΩ to 5Ω.
Measurement of Medium Resistance (1Ω - 100kΩ)Following are the methods employed for measuring a resistance whose value is in the range 1Ω - 100kΩ -
- Ammeter-Voltmeter Method
- Wheatstone Bridge Method
- Substitution Method
- Carey- Foster Bridge Method
- Ohmmeter Method
Ammeter Voltmeter MethodThis is the most crude and simplest method of measuring resistance. It uses one ammeter to measure current, I and one voltmeter to measure voltage, V and we get the value of resistance as Now we can have two possible connections of ammeter and voltmeter, shown in the figure below. Now in figure 1, the voltmeter measures voltage drop across ammeter and the unknown resistance, hence Hence, the relative error will be, For connection in figure 2, the ammeter measures the sum of current through voltmeter and resistance, hence The relative error will be, It can be observed that the relative error is zero for Ra = 0 in first case and Rv = ∞ in second case. Now the questions stand that which connection to be used in which case. To find out this we equate both the errors Hence for resistances greater than that given by above equation we use the first method and for less than that we use second method.
Wheatstone Bridge MethodThis is the simplest and the most basic bridge circuit used in measurement studies. It mainly consists of four arms of resistance P, Q; R and S. R is the unknown resistance under experiment, while S is a standard resistance. P and Q are known as the ratio arms. An EMF source is connected between points a and b while a galvanometer is connected between points c and d. A bridge circuit always works on the principle of null detection, i.e. we vary a parameter until the detector shows zero and then use a mathematical relation to determine the unknown in terms of varying parameter and other constants. Here also the standard resistance, S is varied in order to obtain null deflection in the galvanometer. This null deflection implies no current from point c to d, which implies that potential of point c and d is same. Hence Combining the above two equations we get the famous equation –
Substitution MethodThe figure below shows the circuit diagram for resistance measurement of an unknown resistance R. S is a standard variable resistance and r is a regulating resistance. First the switch is place at position 1 and the ammeter is made to read a certain amount of current by varying r. The value of ammeter reading is noted. Now the switch is moved to position 2 and S is varied in order to achieve the same ammeter reading as it read in the initial case. The value of S for which ammeter reads same as in position 1, is the value of unknown resistance R, provided the EMF source has constant value throughout the experiment.
Measurement of High Resistance (>100kΩ)Following are few methods used for measurement of high resistance values-
- Loss of Charge Method
- Megohm bridge Method
- Direct Deflection Method
- Electrostatic charges can get accumulated on measuring instruments
- Leakage current becomes comparable to measuring current and can cause error
- Insulation resistance is one of the most common in this category; however a dielectric is always modeled as a resistor and capacitor in parallel. Hence while measuring the insulation resistance (I.R.) the current includes both the component and hence true value of resistance is not obtained. The capacitive component though falls exponentially but still takes very long time to decay. Hence different values of I.R. are obtained at different times.
- Protection of delicate instruments from high fields.
Loss of Charge MethodIn this method we utilize the equation of voltage across a discharging capacitor to find the value of unknown resistance R. Figure below shows the circuit diagram and the equations involved are- However the above case assumes no leakage resistance of the capacitor. Hence to account for it we use the circuit shown in the figure below. R1 is the leakage resistance of C and R is the unknown resistance. We follow the same procedure but first with switch S1 closed and next with switch S1 open. For the first case we get For second case with switch open we get Using R1 from above equation in equation for R’ we can find R.
Megohm Bridge MethodIn this method we use the famous Wheatstone bridge philosophy but in a slightly modified way. A high resistance is represented as in the figure below. G is the guard terminal. Now we can also represent the resistor as shown in the adjoining figure, where RAG and RBG are the leakage resistances. The circuit for measurement is shown in the figure below. It can be observed that we actually obtain the resistance which is parallel combination of R and RAG. Although this causes very insignificant error.
MeggerMegger is one of the most important measuring device used by electrical engineers and is essentially used for measuring insulation resistance only. It consists of a generator which can be hand driven or nowadays we have electronic megger. Details of megger have been discussed in a separate article.
Closely Related Articles More Related Articles Digital PotentiometersWheatstone Bridge Circuit Theory and PrincipleKelvin Bridge Circuit | Kelvin Double BridgeMaxwell Bridge Inductance Capacitance BridgeAnderson′s Bridge | Advantages Disadvantages of Anderson′s BridgeHay′s Bridge Circuit Theory Phasor Diagram Advantages ApplicationsOwens Bridge Circuit and AdvantagesSchering Bridge Measurement of Capacitance using Schering BridgeDe Sauty BridgeHeaviside Bridge CircuitBlavier Test | Murray Loop Test | Varley Loop Test | Fisher Loop TestSensor | Types of SensorTemperature Sensor Temperature Measurement | Types of Temperature SensorVoltage SensorAir MeterWater MeterFlow MeasurementFlow MeterDigital Frequency MeterElectrical Measuring Instruments | Types Accuracy Precision Resolution SpeedTransducer | Types of TransducerCathode Ray Oscilloscope | CROOhmmeter Working Principle of OhmmeterErrors in Measurement | Classification of ErrorsLissajous Patterns of CRO or Cathode Ray OscilloscopeFrequency Limitation of an OscilloscopePotentiometer Working Principle of PotentiometerPermanent Magnet Moving Coil Instrument or PMMC InstrumentCharacteristics of SensorsSignal GeneratorMoving Iron InstrumentInduction Type MetersMeasurement of Voltage Current and Frequency by OscilloscopeElectrostatic Type Instruments Construction Principle Torque EquationWhat is Low Power Factor Wattmeter ?Resistance Temperature Detector or RTD | Construction and Working PrincipleEnergy Meter with Lag Adjustment DevicesRectifier Type Instrument | Construction Principle of OperationThermistor Thermometer | Thermistor Temperature Sensor | Construction and Principle Digital Storage OscilloscopeElectrodynamometer Type WattmeterThermocouple type Instruments Construction Principle of OperationThermistor Definition Properties Construction Characteristics and Applications Double Beam OscilloscopeProtection of MetersMeasurement of Three Phase PowerBimetallic Strip ThermometerSampling OscilloscopeVarmeter | Single Phase and Polyphase VarmeterThermocouple Temperature MeasurementTemperature TransducersMegger | Working Principle Types History Uses of MeggerRadiation Pyrometer | Types Working PrincipleOptical Pyrometer | Construction and Working PrinciplePiezoelectric TransducerStrain GaugeWeston Type Frequency MeterPower Factor Meters | Electrodynamometer Type Power Factor MeterInductive TransducersPhase Sequence IndicatorAmmeter Working Principle and Types of AmmeterLinear Variable Differential Transformer LVDTOscillator TransducerDigital MultimeterInsulation Resistance Test and Polarization Index TestHartley OscillatorTan Delta Test |Loss Angle Test | Dissipation Factor TestColpitts OscillatorElectronic DC VoltmeterWorking Principle of Voltmeter and Types of VoltmeterClapp OscillatorRC Phase Shift OscillatorDigital Voltmeters Working Principle of Digital VoltmeterWien Bridge OscillatorMeasurement of Electrical EnergyGunn OscillatorEnergy Meter TestingCrystal OscillatorAC PotentiometerMonostable MultivibratorConstruction of AC Energy MeterBistable MultivibratorWatt Hour MeterCapacitance MeterWhat is an Oscillator?Vector Impedance MeterAstable MultivibratorHow to Use a Digital Multimeter?Voltage Controlled Oscillator | VCONew Articles Acidity Test of Transformer Insulating OilMagnetic FluxRing CounterDischarging a CapacitorCharging a Capacitor