Voltage in Parallel
Voltage in Series
Voltage Regulator 7805
How to Use a Digital Multimeter?Closely Related Articles
Electrical Conductance Conductivity of Metal Semiconductor and Insulator | Band Theory
Properties of Electric Conductor
Electrical Resistance and Laws of Resistance
Measurement of Resistance
Questions on Capacitor
Questions on Magnet Field
Electric Current and Theory of Electricity | Heating and Magnetic Effect
Nature of Electricity
Drift Velocity Drift Current and Electron Mobility
Electric Current and Voltage Division Rule
RMS or Root Mean Square Value of AC Signal
Voltage or Electric Potential Difference
Safety Rules for Medium Voltage Installation
Voltage Drop Calculation
Voltage Regulator 7805
Voltage in Series
Voltage in Parallel
Electric Circuit and Electrical Circuit Elements
Series Parallel Battery Cells
Electrical DC Series and Parallel Circuit
RL Series Circuit
Three Phase Circuit | Star and Delta System
RL Parallel Circuit
RL Circuit Transfer Function Time Constant RL Circuit as Filter
Construction of AC Circuits and Working of AC Circuits
Series RLC Circuit
Parallel RLC Circuit
Resistances in Series and Resistances in Parallel
Resonance in Series RLC Circuit
Planar and Non Planar Graphs of Circuit
What is Electric Field?
Static Electric Field | Electrostatic Induction | Electric Field Strength
What is Flux? Types of Flux?
Magnetic Field and Magnetic Circuit | Magnetic Materials
Energy Stored in a Magnetic Field
A Current Carrying Conductor Within A Magnetic Field
Hard Magnetic Materials
Soft Magnetic Materials
Magnetic Circuit with Air Gap
Electrical International Symbol
Types of resistor Carbon Composition and Wire Wound Resistor
Varistor Metal Oxide Varistor is nonlinear Resistor
Carbon Composition Resistor
Wire Wound Resistor
Variable Resistors | Defination, Uses and Types of Variable Resistors
Light Dependent Resistor | LDR and Working Principle of LDR
What is Capacitor and Capacitance? Types of Capacitors
Working Principle of a Capacitor
Energy Stored in Capacitor
Quality Factor of Inductor and Capacitor
Transient Behavior of Capacitor
Capacitors in Series and Parallel
Testing of Capacitor Bank
How to Test Capacitors?
Join us as Electrical Content Writer
Questions on Battery
Questions on Electrical Earthing
SI System of Units
Analysis of Exponential Fourier Series
What is Inductor and Inductance | Theory of Inductor
Applications of Shift Registers
Trigonometric Fourier Series
Fourier Series and Fourier Transform
Questions on Alternating Current
Questions on Insulator
Questions on Electrolysis
Questions on Electrical Fuse
Questions on Current and Voltage Source
Questions on Magnetic Materials
Electric Power Single and Three Phase Power Active Reactive Apparent
Vector Algebra | Vector Diagram
Relationship of Line and Phase Voltages and Currents in a Star Connected System
Vector Diagram | Three Phase Vector Diagram
Source of Electrical Energy
Earn with us
Measurement of Resistance
Measurement of Low Resistance (<1Ω)The major problem in measurement of low resistance values is the contact resistance or lead resistance of the measuring instruments, though being small in value is comparable to the resistance being measured and hence causes serious error.
Thus to eliminate this issue small valued resistance are constructed with four terminals. Two terminals are current terminals and other two are potential terminals. Figure below shows the construction of low resistance. The current is flown through current terminals C1 and C2 while the potential drop is measured across potential terminals V1 and V2. Hence we can find out the value of resistance under experiment in terms of V and I as indicated in the above figure. This method helps us to exclude the contact resistance due to current terminals and though contact resistance of potential terminals still comes into picture, it is very small fraction of high resistance potential circuit and hence induces negligible error. The methods employed for measurement of low resistances are:-
- Kelvin’s Double Bridge Method
- Potentiometer Method
- Ducter Ohmmeter.
Kelvin’s Double BridgeKelvin’s double bridge is a modification of simple Wheatstone bridge. Figure below shows the circuit diagram of Kelvin’s double bridge. As we can see in the above figure there are two sets of arms, one with resistances P and Q and other with resistances p and q. R is the unknown low resistance and S is a standard resistance. Here r represents the contact resistance between the unknown resistance and the standard resistance, whose effect we need to eliminate. For measurement we make the ratio P/Q equal to p/q and hence a balanced Wheatstone bridge is formed leading to null deflection in the galvanometer. Hence for a balanced bridge we can write Putting eqn 2 in 1 and solving and using P/Q = p/q, we get- Hence we see that by using balanced double arms we can eliminate the contact resistance completely and hence error due to it. To eliminate another error caused due to thermo-electric emf, we take another reading with battery connection reversed and finally take average of the two readings. This bridge is useful for resistances in range of 0.1µΩ to 1.0 Ω.
Ducter OhmmeterIt is an electromechanical instrument used for measurement of low resistances. It comprises of a permanent magnet similar to that of a PMMC instrument and two coils in between the magnetic field created by the poles of the magnet. The two coils are at right angles to each other and are free to rotate about the common axis. Figure below shows a Ducter Ohmmeter and the connections required to measure an unknown resistance R. One of the coil called current coil, is connected to current terminals C1 and C2, while the other coil called, voltage coil is connected to potential terminals V1 and V2. Voltage coil carries current proportional of the voltage drop across R and so is its torque produced. Current coil carries current proportional to the current flowing through R and so is its torque too. Both the torque acts in opposite direction and the indicator come to halt when the two are equal. This instrument is useful for resistance in range 100µΩ to 5Ω.
Measurement of Medium Resistance (1Ω - 100kΩ)Following are the methods employed for measuring a resistance whose value is in the range 1Ω - 100kΩ -
- Ammeter- Voltmeter Method
- Wheatstone Bridge Method
- Substitution Method
- Carey- Foster Bridge Method
- Ohmmeter Method
Ammeter Voltmeter MethodThis is the most crude and simplest method of measuring resistance. It uses one ammeter to measure current, I and one voltmeter to measure voltage, V and we get the value of resistance as Now we can have two possible connections of ammeter and voltmeter, shown in the figure below. Now in figure 1, the voltmeter measures voltage drop across ammeter and the unknown resistance, hence Hence, the relative error will be, For connection in figure 2, the ammeter measures the sum of current through voltmeter and resistance, hence The relative error will be, It can be observed that the relative error is zero for Ra = 0 in first case and Rv = ∞ in second case. Now the questions stand that which connection to be used in which case. To find out this we equate both the errors Hence for resistances greater than that given by above equation we use the first method and for less than that we use second method.
Wheatstone Bridge MethodThis is the simplest and the most basic bridge circuit used in measurement studies. It mainly consists of four arms of resistance P, Q; R and S. R is the unknown resistance under experiment, while S is a standard resistance. P and Q are known as the ratio arms. An EMF source is connected between points a and b while a galvanometer is connected between points c and d. A bridge circuit always works on the principle of null detection, i.e. we vary a parameter until the detector shows zero and then use a mathematical relation to determine the unknown in terms of varying parameter and other constants. Here also the standard resistance, S is varied in order to obtain null deflection in the galvanometer. This null deflection implies no current from point c to d, which implies that potential of point c and d is same. Hence Combining the above two equations we get the famous equation –
Substitution MethodThe figure below shows the circuit diagram for resistance measurement of an unknown resistance R. S is a standard variable resistance and r is a regulating resistance. First the switch is place at position 1 and the ammeter is made to read a certain amount of current by varying r. The value of ammeter reading is noted. Now the switch is moved to position 2 and S is varied in order to achieve the same ammeter reading as it read in the initial case. The value of S for which ammeter reads same as in position 1, is the value of unknown resistance R, provided the EMF source has constant value throughout the experiment.
Measurement of High Resistance (>100kΩ)Following are few methods used for measurement of high resistance values-
- Loss of Charge Method
- Megohm bridge Method
- Direct Deflection Method
- Electrostatic charges can get accumulated on measuring instruments
- Leakage current becomes comparable to measuring current and can cause error
- Insulation resistance is one of the most common in this category; however a dielectric is always modeled as a resistor and capacitor in parallel. Hence while measuring the insulation resistance (I.R.) the current includes both the component and hence true value of resistance is not obtained. The capacitive component though falls exponentially but still takes very long time to decay. Hence different values of I.R. are obtained at different times.
- Protection of delicate instruments from high fields.
Loss of Charge MethodIn this method we utilize the equation of voltage across a discharging capacitor to find the value of unknown resistance R. Figure below shows the circuit diagram and the equations involved are- However the above case assumes no leakage resistance of the capacitor. Hence to account for it we use the circuit shown in the figure below. R1 is the leakage resistance of C and R is the unknown resistance. We follow the same procedure but first with switch S1 closed and next with switch S1 open. For the first case we get For second case with switch open we get Using R1 from above equation in equation for R’ we can find R.
Megohm Bridge MethodIn this method we use the famous Wheatstone bridge philosophy but in a slightly modified way. A high resistance is represented as in the figure below. G is the guard terminal. Now we can also represent the resistor as shown in the adjoining figure, where RAG and RBG are the leakage resistances. The circuit for measurement is shown in the figure below. It can be observed that we actually obtain the resistance which is parallel combination of R and RAG. Although this causes very insignificant error.
MeggerMegger is one of the most important measuring device used by electrical engineers and is essentially used for measuring insulation resistance only. It consists of a generator which can be hand driven or nowadays we have electronic megger. Details of megger have been discussed in a separate article.
Closely Related Articles Electrical Conductance Conductivity of Metal Semiconductor and Insulator | Band TheoryProperties of Electric ConductorElectrical Resistance and Laws of ResistanceSeries ResistanceMore Related Articles Questions on CapacitorQuestions on Magnet FieldElectric Current and Theory of Electricity | Heating and Magnetic EffectNature of ElectricityDrift Velocity Drift Current and Electron MobilityElectric Current and Voltage Division RuleRMS or Root Mean Square Value of AC SignalVoltage or Electric Potential DifferenceVoltage DividerSafety Rules for Medium Voltage InstallationVoltage MultiplierVoltage Drop CalculationVoltage DoublerVoltage RegulatorVoltage SourceVoltage FollowerVoltage Regulator 7805Voltage in SeriesVoltage in ParallelVoltage SensorElectric Circuit and Electrical Circuit ElementsSeries Parallel Battery CellsElectrical DC Series and Parallel CircuitRL Series CircuitRLC CircuitThree Phase Circuit | Star and Delta SystemRL Parallel CircuitRL Circuit Transfer Function Time Constant RL Circuit as FilterConstruction of AC Circuits and Working of AC CircuitsSeries RLC CircuitParallel RLC CircuitResistances in Series and Resistances in ParallelResonance in Series RLC CircuitPlanar and Non Planar Graphs of CircuitClipping CircuitWhat is Electric Field?Static Electric Field | Electrostatic Induction | Electric Field StrengthWhat is Flux? Types of Flux?Magnetic PermeabilityMagnetic Field and Magnetic Circuit | Magnetic MaterialsMagnetic SaturationEnergy Stored in a Magnetic FieldHysteresis LoopA Current Carrying Conductor Within A Magnetic FieldMagnetic SusceptibilityHard Magnetic MaterialsSoft Magnetic MaterialsMagnetic Circuit with Air GapElectrical International SymbolTypes of resistor Carbon Composition and Wire Wound ResistorVaristor Metal Oxide Varistor is nonlinear ResistorCarbon Composition ResistorWire Wound ResistorVariable Resistors | Defination, Uses and Types of Variable ResistorsLight Dependent Resistor | LDR and Working Principle of LDRWhat is Capacitor and Capacitance? Types of CapacitorsWorking Principle of a CapacitorEnergy Stored in CapacitorQuality Factor of Inductor and CapacitorTransient Behavior of CapacitorCylindrical CapacitorSpherical CapacitorCapacitors in Series and ParallelTesting of Capacitor BankHow to Test Capacitors?Join us as Electrical Content Writer Questions on BatteryQuestions on Electrical EarthingSI System of UnitsAnalysis of Exponential Fourier SeriesWhat is Inductor and Inductance | Theory of InductorApplications of Shift RegistersMutual InductanceSelf InductanceTrigonometric Fourier SeriesFourier Series and Fourier TransformQuestions on Alternating CurrentElectric HeatingQuestions on InsulatorQuestions on ElectrolysisQuestions on Electrical FuseQuestions on Current and Voltage SourceQuestions on Magnetic MaterialsElectric Power Single and Three Phase Power Active Reactive ApparentVector Algebra | Vector DiagramRelationship of Line and Phase Voltages and Currents in a Star Connected SystemVector Diagram | Three Phase Vector DiagramSource of Electrical EnergyEarn with usNew Articles Voltage SensorFlow MeasurementVoltage in ParallelVoltage in SeriesVoltage Regulator 7805How to Use a Digital Multimeter?