# Wheatstone Bridge Circuit Theory and Principle

## Wheatstone Bridge

For measuring accurately any electrical resistance**Wheatstone bridge**is widely used. There are two known resistors, one variable resistor and one unknown resistor connected in bridge form as shown below. By adjusting the variable resistor the current through the Galvanometer is made zero. When the current through the galvanometer becomes zero, the ratio of two known resistors is exactly equal to the ratio of adjusted value of variable resistance and the value of unknown resistance. In this way the value of unknown electrical resistance can easily be measured by using a Wheatstone Bridge.

## Wheatstone Bridge Theory

The general arrangement of**Wheatstone bridge circuit**is shown in the figure below. It is a four arms bridge circuit where arm AB, BC, CD and AD are consisting of electrical resistances P, Q, S and R respectively. Among these resistances P and Q are known fixed electrical resistances and these two arms are referred as ratio arms. An accurate and sensitive Galvanometer is connected between the terminals B and D through a switch S

_{2}.

The voltage source of this

**Wheatstone bridge**is connected to the terminals A and C via a switch S

_{1}as shown. A variable resistor S is connected between point C and D. The potential at point D can be varied by adjusting the value of variable resistor. Suppose current I

_{1}and current I

_{2}are flowing through the paths ABC and ADC respectively. If we vary the electrical resistance value of arm CD the value of current I

_{2}will also be varied as the voltage across A and C is fixed. If we continue to adjust the variable resistance one situation may comes when voltage drop across the resistor S that is I

_{2}. S is becomes exactly equal to voltage drop across resistor Q that is I

_{1}.Q. Thus the potential at point B becomes equal to the potential at point D hence potential difference between these two points is zero hence current through galvanometer is nil. Then the deflection in the galvanometer is nil when the switch S

_{2}is closed.

Now, from

**Wheatstone bridge circuit**and Now potential of point B in respect of point C is nothing but the voltage drop across the resistor Q and this is Again potential of point D in respect of point C is nothing but the voltage drop across the resistor S and this is Equating, equations (i) and (ii) we get, Here in the above equation, the value of S and P ⁄ Q are known, so value of R can easily be determined.

The electrical resistances P and Q of the

**Wheatstone bridge**are made of definite ratio such as 1:1; 10:1 or 100:1 known as ratio arms and S the rheostat arm is made continuously variable from 1 to 1,000 Ω or from 1 to 10,000 Ω.

The above explanation is most basic

**Wheatstone bridge theory**.

### Video Presentation of Wheatstone Bridge Theory

Closely Related Articles Kelvin Bridge Circuit | Kelvin Double BridgeMaxwell Bridge Inductance Capacitance BridgeAnderson′s Bridge | Advantages Disadvantages of Anderson′s BridgeHay′s Bridge Circuit Theory Phasor Diagram Advantages ApplicationsOwens Bridge Circuit and AdvantagesSchering Bridge Measurement of Capacitance using Schering BridgeDe Sauty BridgeHeaviside Bridge CircuitBlavier Test | Murray Loop Test | Varley Loop Test | Fisher Loop TestMore Related Articles Digital PotentiometersSensor | Types of SensorTemperature Sensor Temperature Measurement | Types of Temperature SensorVoltage SensorMeasurement of ResistanceAir MeterWater MeterWatt Hour MeterCapacitance MeterVector Impedance MeterHow to Use a Digital Multimeter?Flow MeasurementFlow MeterDigital Frequency MeterElectrical Measuring Instruments | Types Accuracy Precision Resolution SpeedTransducer | Types of TransducerCathode Ray Oscilloscope | CROOhmmeter Working Principle of OhmmeterErrors in Measurement | Classification of ErrorsLissajous Patterns of CRO or Cathode Ray OscilloscopeFrequency Limitation of an OscilloscopePotentiometer Working Principle of PotentiometerPermanent Magnet Moving Coil Instrument or PMMC InstrumentCharacteristics of SensorsSignal GeneratorMoving Iron InstrumentInduction Type MetersMeasurement of Voltage Current and Frequency by OscilloscopeElectrostatic Type Instruments Construction Principle Torque EquationWhat is Low Power Factor Wattmeter ?Resistance Temperature Detector or RTD | Construction and Working PrincipleEnergy Meter with Lag Adjustment DevicesRectifier Type Instrument | Construction Principle of OperationThermistor Thermometer | Thermistor Temperature Sensor | Construction and Principle Digital Storage OscilloscopeElectrodynamometer Type WattmeterThermocouple type Instruments Construction Principle of OperationThermistor Definition Properties Construction Characteristics and Applications Double Beam OscilloscopeProtection of MetersMeasurement of Three Phase PowerBimetallic Strip ThermometerSampling OscilloscopeVarmeter | Single Phase and Polyphase VarmeterThermocouple Temperature MeasurementTemperature TransducersMegger | Working Principle Types History Uses of MeggerRadiation Pyrometer | Types Working PrincipleOptical Pyrometer | Construction and Working PrinciplePiezoelectric TransducerStrain GaugeWeston Type Frequency MeterPower Factor Meters | Electrodynamometer Type Power Factor MeterInductive TransducersPhase Sequence IndicatorAmmeter Working Principle and Types of AmmeterLinear Variable Differential Transformer LVDTOscillator TransducerDigital MultimeterInsulation Resistance Test and Polarization Index TestHartley OscillatorTan Delta Test |Loss Angle Test | Dissipation Factor TestColpitts OscillatorElectronic DC VoltmeterWorking Principle of Voltmeter and Types of VoltmeterClapp OscillatorRC Phase Shift OscillatorDigital Voltmeters Working Principle of Digital VoltmeterWien Bridge OscillatorMeasurement of Electrical EnergyGunn OscillatorEnergy Meter TestingCrystal OscillatorAC PotentiometerMonostable MultivibratorConstruction of AC Energy MeterBistable MultivibratorWhat is an Oscillator?Astable MultivibratorVoltage Controlled Oscillator | VCONew Articles Ring CounterDischarging a CapacitorCharging a CapacitorElectric PotentialParity Generator