# Maxwell Bridge Inductance Capacitance Bridge

**Maxwell bridge**is an AC bridge so before going in further detail let us know more about the AC bridge.

## AC Bridges

AC Bridges consist of a source, balance detector and four arms. In AC bridges, all the four arms consists of impedance. The AC bridges are formed by replacing the DC battery with an AC source and galvanometer by detector of Wheatstone bridge.They are highly useful to find out inductance, capacitance, storage factor, dissipation factor etc.

Now let us derive general expression for an AC bridge balance.

Figure given below shows AC bridge network:
Here Z_{1}, Z_{2}, Z_{3} and Z_{4} are the arms of the bridge.

Now at the balance condition, the potential difference between b and d must be zero. From this, when the voltage drop from a to d equals to drop from a to b both in magnitude and phase.

Thus, we have from figure e_{1} = e_{2}
From equation 1, 2 and 3 we have Z_{1}.Z_{4} = Z_{2}.Z_{3} and when impedance are replaced by admittance, we have Y_{1}.Y_{4} = Y_{2}.Y_{3}.

Now consider the basic form of an AC bridge. Suppose we have bridge circuit as shown below,
In this circuit R_{3} and R_{4} are pure electrical resistances. Putting the value of Z_{1}, Z_{2}, Z_{3} and Z_{4} in the equation that we have derived above for AC bridge.
Now equating the real and imaginary parts we get
Following are the important conclusions that can be drawn from the above equations:

- We get two balanced equations that are obtained by equating real and imaginary parts this means that for an ac bridge both the relation (i.e.magnitude and phase) must be satisfied at the same time. Both the equations are said to be independent if and only if both equation contain single variable element. This variable can be inductor or resistor.
- The above equations are independent of frequency that means we do not require exact frequency of the source voltage and also the applied source voltage waveform need not to be perfectly sinusoidal.

## Maxwell's Bridge

Under this we going to study about the following**Maxwell's inductor bridge****Maxwell's inductor capacitance bridge**

### Maxwell's Inductance Bridge

Let us now discuss**Maxwell's inductor bridge**. The figure shows the circuit diagram of Maxwell's inductor bridge. In this bridge the arms bc and cd are purely resistive while the phase balance depends on the arms ab and ad.

Here l

_{1}= unknown inductor of r

_{1}.

l

_{2}= variable inductor of resistance R

_{2}.

r

_{2}= variable electrical resistance.

As we have discussed in AC bridge according to balance condition, we have at balance point We can vary R

_{3}and R

_{4}from 10 ohms to 10,000 ohms with the help of resistance box.

### Maxwell's Inductance Capacitance Bridge

In this**Maxwell Bridge**, the unknown inductor is measured by the standard variable capacitor.

Circuit of this bridge is given below, Here, l

_{1}is unknown inductance, C

_{4}is standard capacitor. Now under balance conditions we have from ac bridge that Z

_{1}.Z

_{4}= Z

_{2}.Z

_{3}Let us separate the real and imaginary parts, the we have, Now the quality factor is given by,

### Advantages of Maxwell's Bridge

Advantages of Maxwell's bridge are showing below- The frequency does not appear in the final expression of both equations, hence it is independent of frequency.
**Maxwell's inductor capacitance bridge**is very useful for the wide range of measurement of inductor at audio frequencies.

### Disadvantages of Maxwell's Bridge

- The variable standard capacitor is very expensive.
- The bridge is limited to measurement of low quality coils (1 < Q < 10) and it is also unsuitable for low value of Q (i.e. Q < 1) from this we conclude that a Maxwell bridge is used suitable only for medium Q coils.

Closely Related Articles Wheatstone Bridge Circuit Theory and PrincipleKelvin Bridge Circuit | Kelvin Double BridgeSensor | Types of SensorTemperature Sensor Temperature Measurement | Types of Temperature SensorAnderson′s Bridge | Advantages Disadvantages of Anderson′s BridgeHay′s Bridge Circuit Theory Phasor Diagram Advantages ApplicationsOwens Bridge Circuit and AdvantagesSchering Bridge Measurement of Capacitance using Schering BridgeDe Sauty BridgeHeaviside Bridge CircuitBlavier Test | Murray Loop Test | Varley Loop Test | Fisher Loop TestMore Related Articles Cathode Ray Oscilloscope | CROLissajous Patterns of CRO or Cathode Ray OscilloscopeMeasurement of Voltage Current and Frequency by OscilloscopeDigital Storage OscilloscopeDouble Beam OscilloscopeSampling OscilloscopeMeasurement of ResistanceDigital Frequency MeterOhmmeter Working Principle of OhmmeterInduction Type MetersWhat is Low Power Factor Wattmeter ?Electrodynamometer Type WattmeterProtection of MetersBimetallic Strip ThermometerVarmeter | Single Phase and Polyphase VarmeterWeston Type Frequency MeterAmmeter Working Principle and Types of AmmeterDigital MultimeterElectronic DC VoltmeterWorking Principle of Voltmeter and Types of VoltmeterDigital Voltmeters Working Principle of Digital VoltmeterConstruction of AC Energy MeterWatt Hour MeterCapacitance MeterVector Impedance MeterHow to Use a Digital Multimeter?Flow MeasurementFlow MeterAir MeterWater MeterFrequency Limitation of an OscilloscopeSignal GeneratorHartley OscillatorColpitts OscillatorClapp OscillatorRC Phase Shift OscillatorWien Bridge OscillatorGunn OscillatorCrystal OscillatorWhat is an Oscillator?Voltage Controlled Oscillator | VCOPotentiometer Working Principle of PotentiometerAC PotentiometerDigital PotentiometersElectrical Measuring Instruments | Types Accuracy Precision Resolution SpeedTransducer | Types of TransducerErrors in Measurement | Classification of ErrorsPermanent Magnet Moving Coil Instrument or PMMC InstrumentCharacteristics of SensorsMoving Iron InstrumentElectrostatic Type Instruments Construction Principle Torque EquationResistance Temperature Detector or RTD | Construction and Working PrincipleEnergy Meter with Lag Adjustment DevicesRectifier Type Instrument | Construction Principle of OperationThermistor Thermometer | Thermistor Temperature Sensor | Construction and Principle Thermocouple Type Instruments Construction Principle of OperationThermistor Definition Properties Construction Characteristics and Applications Measurement of Three Phase PowerThermocouple Temperature MeasurementTemperature TransducersMegger | Working Principle Types History Uses of MeggerRadiation Pyrometer | Types Working PrincipleOptical Pyrometer | Construction and Working PrinciplePiezoelectric TransducerStrain GaugePower Factor Meters | Electrodynamometer Type Power Factor MeterInductive TransducersPhase Sequence IndicatorLinear Variable Differential Transformer LVDTOscillator TransducerPolarization Index Test or PI TestVoltage SensorTan Delta Test |Loss Angle Test | Dissipation Factor TestMeasurement of Electrical EnergyEnergy Meter TestingMonostable MultivibratorBistable MultivibratorAstable MultivibratorNew Articles Series and Parallel Inductors Electric PowerMeasurement of Losses in Shunt ReactorThree Phase Shunt ReactorMeasurement of Insulation ResistanceAmpere's Circuital Law