Acidity Test of Transformer Insulating Oil
Discharging a Capacitor
Charging a CapacitorClosely Related Articles
Wheatstone Bridge Circuit Theory and Principle
Kelvin Bridge Circuit | Kelvin Double Bridge
Maxwell Bridge Inductance Capacitance Bridge
Anderson′s Bridge | Advantages Disadvantages of Anderson′s Bridge
Hay′s Bridge Circuit Theory Phasor Diagram Advantages Applications
Owens Bridge Circuit and Advantages
Schering Bridge Measurement of Capacitance using Schering Bridge
De Sauty Bridge
Heaviside Bridge Circuit
Blavier Test | Murray Loop Test | Varley Loop Test | Fisher Loop Test
Sensor | Types of Sensor
Temperature Sensor Temperature Measurement | Types of Temperature Sensor
Measurement of Resistance
Digital Frequency Meter
Electrical Measuring Instruments | Types Accuracy Precision Resolution Speed
Cathode Ray Oscilloscope | CRO
Errors in Measurement | Classification of Errors
Lissajous Patterns of CRO or Cathode Ray Oscilloscope
Frequency Limitation of an Oscilloscope
Permanent Magnet Moving Coil Instrument or PMMC Instrument
Characteristics of Sensors
Moving Iron Instrument
Induction Type Meters
Measurement of Voltage Current and Frequency by Oscilloscope
Electrostatic Type Instruments Construction Principle Torque Equation
What is Low Power Factor Wattmeter ?
Resistance Temperature Detector or RTD | Construction and Working Principle
Energy Meter with Lag Adjustment Devices
Rectifier Type Instrument | Construction Principle of Operation
Digital Storage Oscilloscope
Electrodynamometer Type Wattmeter
Thermocouple type Instruments Construction Principle of Operation
Double Beam Oscilloscope
Protection of Meters
Measurement of Three Phase Power
Bimetallic Strip Thermometer
Varmeter | Single Phase and Polyphase Varmeter
Thermocouple Temperature Measurement
Megger | Working Principle Types History Uses of Megger
Radiation Pyrometer | Types Working Principle
Weston Type Frequency Meter
Power Factor Meters | Electrodynamometer Type Power Factor Meter
Phase Sequence Indicator
Ammeter Working Principle and Types of Ammeter
Insulation Resistance Test and Polarization Index Test
Tan Delta Test |Loss Angle Test | Dissipation Factor Test
Electronic DC Voltmeter
Working Principle of Voltmeter and Types of Voltmeter
RC Phase Shift Oscillator
Digital Voltmeters Working Principle of Digital Voltmeter
Wien Bridge Oscillator
Measurement of Electrical Energy
Energy Meter Testing
Construction of AC Energy Meter
Watt Hour Meter
Vector Impedance Meter
How to Use a Digital Multimeter?
Voltage Controlled Oscillator
Maxwell Bridge Inductance Capacitance Bridge
AC BridgesAC Bridges consist of a source, balance detector and four arms. In AC bridges, all the four arms consists of impedance. The AC bridges are formed by replacing the DC battery with an AC source and galvanometer by detector of Wheatstone bridge.
They are highly useful to find out inductance, capacitance, storage factor, dissipation factor etc. Now let us derive general expression for an AC bridge balance. Figure given below shows AC bridge network: Here Z1, Z2, Z3 and Z4 are the arms of the bridge. Now at the balance condition, the potential difference between b and d must be zero. From this, when the voltage drop from a to d equals to drop from a to b both in magnitude and phase.
Thus, we have from figure e1 = e2 From equation 1, 2 and 3 we have Z1.Z4 = Z2.Z3 and when impedance are replaced by admittance, we have Y1.Y4 = Y2.Y3. Now consider the basic form of an AC bridge. Suppose we have bridge circuit as shown below, In this circuit R3 and R4 are pure electrical resistances. Putting the value of Z1, Z2, Z3 and Z4 in the equation that we have derived above for AC bridge. Now equating the real and imaginary parts we get Following are the important conclusions that can be drawn from the above equations:
- We get two balanced equations that are obtained by equating real and imaginary parts this means that for an ac bridge both the relation (i.e.magnitude and phase) must be satisfied at the same time. Both the equations are said to be independent if and only if both equation contain single variable element. This variable can be inductor or resistor.
- The above equations are independent of frequency that means we do not require exact frequency of the source voltage and also the applied source voltage waveform need not to be perfectly sinusoidal.
Maxwell's BridgeUnder this we going to study about the following
- Maxwell's inductor bridge
- Maxwell's inductor capacitance bridge
Maxwell's Inductance BridgeLet us now discuss Maxwell's inductor bridge. The figure shows the circuit diagram of Maxwell's inductor bridge. In this bridge the arms bc and cd are purely resistive while the phase balance depends on the arms ab and ad. Here l1 = unknown inductor of r1. l2 = variable inductor of resistance R2. r2 = variable electrical resistance. As we have discussed in AC bridge according to balance condition, we have at balance point We can vary R3 and R4 from 10 ohms to 10,000 ohms with the help of resistance box.
Maxwell's Inductance Capacitance BridgeIn this Maxwell Bridge, the unknown inductor is measured by the standard variable capacitor. Circuit of this bridge is given below, Here, l1 is unknown inductance, C4 is standard capacitor. Now under balance conditions we have from ac bridge that Z1.Z4 = Z2.Z3 Let us separate the real and imaginary parts, the we have, Now the quality factor is given by,
Advantages of Maxwell's BridgeAdvantages of Maxwell's bridge are showing below
- The frequency does not appear in the final expression of both equations, hence it is independent of frequency.
- Maxwell's inductor capacitance bridge is very useful for the wide range of measurement of inductor at audio frequencies.
Disadvantages of Maxwell's Bridge
- The variable standard capacitor is very expensive.
- The bridge is limited to measurement of low quality coils (1 < Q < 10) and it is also unsuitable for low value of Q (i.e. Q < 1) from this we conclude that a Maxwell bridge is used suitable only for medium Q coils.
Closely Related Articles Wheatstone Bridge Circuit Theory and PrincipleKelvin Bridge Circuit | Kelvin Double BridgeAnderson′s Bridge | Advantages Disadvantages of Anderson′s BridgeHay′s Bridge Circuit Theory Phasor Diagram Advantages ApplicationsOwens Bridge Circuit and AdvantagesSchering Bridge Measurement of Capacitance using Schering BridgeDe Sauty BridgeHeaviside Bridge CircuitBlavier Test | Murray Loop Test | Varley Loop Test | Fisher Loop TestMore Related Articles Digital PotentiometersSensor | Types of SensorTemperature Sensor Temperature Measurement | Types of Temperature SensorVoltage SensorMeasurement of ResistanceAir MeterWater MeterFlow MeasurementFlow MeterDigital Frequency MeterElectrical Measuring Instruments | Types Accuracy Precision Resolution SpeedTransducer | Types of TransducerCathode Ray Oscilloscope | CROOhmmeter Working Principle of OhmmeterErrors in Measurement | Classification of ErrorsLissajous Patterns of CRO or Cathode Ray OscilloscopeFrequency Limitation of an OscilloscopePotentiometer Working Principle of PotentiometerPermanent Magnet Moving Coil Instrument or PMMC InstrumentCharacteristics of SensorsSignal GeneratorMoving Iron InstrumentInduction Type MetersMeasurement of Voltage Current and Frequency by OscilloscopeElectrostatic Type Instruments Construction Principle Torque EquationWhat is Low Power Factor Wattmeter ?Resistance Temperature Detector or RTD | Construction and Working PrincipleEnergy Meter with Lag Adjustment DevicesRectifier Type Instrument | Construction Principle of OperationThermistor Thermometer | Thermistor Temperature Sensor | Construction and Principle Digital Storage OscilloscopeElectrodynamometer Type WattmeterThermocouple type Instruments Construction Principle of OperationThermistor Definition Properties Construction Characteristics and Applications Double Beam OscilloscopeProtection of MetersMeasurement of Three Phase PowerBimetallic Strip ThermometerSampling OscilloscopeVarmeter | Single Phase and Polyphase VarmeterThermocouple Temperature MeasurementTemperature TransducersMegger | Working Principle Types History Uses of MeggerRadiation Pyrometer | Types Working PrincipleOptical Pyrometer | Construction and Working PrinciplePiezoelectric TransducerStrain GaugeWeston Type Frequency MeterPower Factor Meters | Electrodynamometer Type Power Factor MeterInductive TransducersPhase Sequence IndicatorAmmeter Working Principle and Types of AmmeterLinear Variable Differential Transformer LVDTOscillator TransducerDigital MultimeterInsulation Resistance Test and Polarization Index TestHartley OscillatorTan Delta Test |Loss Angle Test | Dissipation Factor TestColpitts OscillatorElectronic DC VoltmeterWorking Principle of Voltmeter and Types of VoltmeterClapp OscillatorRC Phase Shift OscillatorDigital Voltmeters Working Principle of Digital VoltmeterWien Bridge OscillatorMeasurement of Electrical EnergyGunn OscillatorEnergy Meter TestingCrystal OscillatorAC PotentiometerMonostable MultivibratorConstruction of AC Energy MeterBistable MultivibratorWatt Hour MeterCapacitance MeterWhat is an Oscillator?Vector Impedance MeterAstable MultivibratorHow to Use a Digital Multimeter?Voltage Controlled Oscillator | VCONew Articles Acidity Test of Transformer Insulating OilMagnetic FluxRing CounterDischarging a CapacitorCharging a Capacitor