# Electrodynamometer Type Wattmeter

**electrodynamometer wattmeter**, it very essential to know the principle of working of

**electrodynamometer type wattmeter**. Dynamometer type wattmeter works on very simple principle and this principle can be stated as "when any current carrying conductor is placed inside a magnetic field, it experiences a mechanical force and due this mechanical force deflection of conductor takes place".

## Construction and Working Principle of Electrodynamometer Type Wattmeter

Now let us look at constructional details of**electrodynamometer**. It consists of following parts There are two types of coils present in the electrodynamometer. They are :

Moving Coil
Moving coil moves the pointer with the help of spring control instrument. A limited amount of current flows through the moving coil so as to avoid heating. So in order to limit the current we have connect the high value resistor in series with the moving coil. The moving is air cored and is mounted on a pivoted spindle and can moves freely. In **electrodynamometer type wattmeter**, moving coil works as pressure coil. Hence moving coil is connected across the voltage and thus the current flowing through this coil is always proportional to the voltage.
Fixed Coil
The fixed coil is divided into two equal parts and these are connected in series with the load, therefore the load current will flow through these coils. Now the reason is very obvious of using two fixed coils instead of one, so that it can be constructed to carry considerable amount of electric current. These coils are called the current coils of **electrodynamometer type wattmeter**. Earlier these fixed coils are designed to carry the current of about 100 amperes but now the modern wattmeter are designed to carry current of about 20 amperes in order to save power.
Control System
Out of two controlling systems i.e.

- Gravity control
- Spring control, only spring controlled systems are used in these types of wattmeter. Gravity controlled system cannot be employed because they will appreciable amount of errors.

Now let us derive the expressions for the controlling torque and deflecting torques. In order to derive these expressions let us consider the circuit diagram given below:
We know that instantaneous torque in electrodynamic type instruments is directly proportional to product of instantaneous values of currents flowing through both the coils and the rate of change of flux linked with the circuit.
Let I_{1} and I_{2} be the instantaneous values of currents in pressure and current coils respectively. So the expression for the torque can be written as:

_{2}= v / R

_{p}where R

_{p}is the resistance of pressure coil.

_{c}= K

_{x}where K is spring constant and x is final steady state value of deflection.

### Advantages of Electrodynamometer Type Wattmeter

Following are the advantages of electrodynamometer type wattmeters and they are written as follows:- Scale is uniform upto certain limit.
- They can be used for both to measure ac as well dc quantities as scale is calibrated for both.

### Errors in Electrodynamometer Type Wattmeter

Following are the errors in the electrodynamometer type wattmeters:- Errors in the pressure coil inductance.
- Errors may be due to pressure coil capacitance.
- Errors may be due to mutual inductance effects.
- Errors may be due connections.(i.e. pressure coil is connected after current coil)
- Error due to Eddy currents.
- Errors caused by vibration of moving system.
- Temperature error.
- Errors due to stray magnetic field.

Closely Related Articles Digital Frequency MeterOhmmeter Working Principle of OhmmeterPotentiometer Working Principle of PotentiometerInduction Type MetersWhat is Low Power Factor Wattmeter ?Energy Meter with Lag Adjustment DevicesMeasurement of Three Phase PowerVarmeter | Single Phase and Polyphase VarmeterMegger | Working Principle Types History Uses of MeggerWeston Type Frequency MeterPower Factor Meters | Electrodynamometer Type Power Factor MeterPhase Sequence IndicatorAmmeter Working Principle and Types of AmmeterDigital MultimeterInsulation Resistance Test and Polarization Index TestTan Delta Test |Loss Angle Test | Dissipation Factor TestElectronic DC VoltmeterWorking Principle of Voltmeter and Types of VoltmeterDigital Voltmeters Working Principle of Digital VoltmeterMeasurement of Electrical EnergyEnergy Meter TestingAC PotentiometerConstruction of AC Energy MeterWatt Hour MeterCapacitance MeterVector Impedance MeterHow to Use a Digital Multimeter?More Related Articles Electrical Measuring Instruments | Types Accuracy Precision Resolution SpeedErrors in Measurement | Classification of ErrorsPermanent Magnet Moving Coil Instrument or PMMC InstrumentMoving Iron InstrumentElectrostatic Type Instruments Construction Principle Torque EquationRectifier Type Instrument | Construction Principle of OperationThermocouple type Instruments Construction Principle of OperationProtection of MetersWheatstone Bridge Circuit Theory and PrincipleKelvin Bridge Circuit | Kelvin Double BridgeMaxwell Bridge Inductance Capacitance BridgeAnderson′s Bridge | Advantages Disadvantages of Anderson′s BridgeHay′s Bridge Circuit Theory Phasor Diagram Advantages ApplicationsOwens Bridge Circuit and AdvantagesSchering Bridge Measurement of Capacitance using Schering BridgeDe Sauty BridgeHeaviside Bridge CircuitBlavier Test | Murray Loop Test | Varley Loop Test | Fisher Loop TestTransducer | Types of TransducerRadiation Pyrometer | Types Working PrincipleResistance Temperature Detector or RTD | Construction and Working PrincipleStrain GaugeTemperature Sensor Temperature Measurement | Types of Temperature SensorSensor | Types of SensorOptical Pyrometer | Construction and Working PrincipleBimetallic Strip ThermometerPiezoelectric TransducerThermistor Thermometer | Thermistor Temperature Sensor | Construction and Principle Thermocouple Temperature MeasurementInductive TransducersTemperature TransducersLinear Variable Differential Transformer LVDTOscillator TransducerThermistor Definition Properties Construction Characteristics and Applications Hartley OscillatorColpitts OscillatorClapp OscillatorRC Phase Shift OscillatorWien Bridge OscillatorGunn OscillatorCrystal OscillatorMonostable MultivibratorBistable Multivibrator555 Timer and 555 Timer WorkingWhat is an Oscillator?Astable MultivibratorVoltage Controlled Oscillator | VCOCathode Ray Oscilloscope | CROLissajous Patterns of CRO or Cathode Ray OscilloscopeFrequency Limitation of an OscilloscopeSignal GeneratorMeasurement of Voltage Current and Frequency by OscilloscopeDigital Storage OscilloscopeDouble Beam OscilloscopeSampling OscilloscopeNew Articles Voltage in ParallelVoltage in SeriesVoltage Regulator 7805How to Use a Digital Multimeter?Vector Impedance MeterVoltage Follower