Acidity Test of Transformer Insulating Oil
Discharging a Capacitor
Charging a CapacitorClosely Related Articles
Digital Frequency Meter
Electrical Measuring Instruments | Types Accuracy Precision Resolution Speed
Cathode Ray Oscilloscope | CRO
Errors in Measurement | Classification of Errors
Lissajous Patterns of CRO or Cathode Ray Oscilloscope
Frequency Limitation of an Oscilloscope
Permanent Magnet Moving Coil Instrument or PMMC Instrument
Characteristics of Sensors
Moving Iron Instrument
Induction Type Meters
Measurement of Voltage Current and Frequency by Oscilloscope
Electrostatic Type Instruments Construction Principle Torque Equation
What is Low Power Factor Wattmeter ?
Resistance Temperature Detector or RTD | Construction and Working Principle
Energy Meter with Lag Adjustment Devices
Rectifier Type Instrument | Construction Principle of Operation
Digital Storage Oscilloscope
Electrodynamometer Type Wattmeter
Thermocouple type Instruments Construction Principle of Operation
Double Beam Oscilloscope
Protection of Meters
Measurement of Three Phase Power
Bimetallic Strip Thermometer
Varmeter | Single Phase and Polyphase Varmeter
Thermocouple Temperature Measurement
Megger | Working Principle Types History Uses of Megger
Radiation Pyrometer | Types Working Principle
Weston Type Frequency Meter
Power Factor Meters | Electrodynamometer Type Power Factor Meter
Phase Sequence Indicator
Ammeter Working Principle and Types of Ammeter
Insulation Resistance Test and Polarization Index Test
Tan Delta Test |Loss Angle Test | Dissipation Factor Test
Electronic DC Voltmeter
Working Principle of Voltmeter and Types of Voltmeter
RC Phase Shift Oscillator
Digital Voltmeters Working Principle of Digital Voltmeter
Wien Bridge Oscillator
Measurement of Electrical Energy
Energy Meter Testing
Construction of AC Energy Meter
Watt Hour Meter
Vector Impedance Meter
How to Use a Digital Multimeter?
Voltage Controlled Oscillator
Wheatstone Bridge Circuit Theory and Principle
Kelvin Bridge Circuit | Kelvin Double Bridge
Maxwell Bridge Inductance Capacitance Bridge
Anderson′s Bridge | Advantages Disadvantages of Anderson′s Bridge
Hay′s Bridge Circuit Theory Phasor Diagram Advantages Applications
Owens Bridge Circuit and Advantages
Schering Bridge Measurement of Capacitance using Schering Bridge
De Sauty Bridge
Heaviside Bridge Circuit
Blavier Test | Murray Loop Test | Varley Loop Test | Fisher Loop Test
Sensor | Types of Sensor
Temperature Sensor Temperature Measurement | Types of Temperature Sensor
Measurement of Resistance
Electrodynamometer Type Wattmeter
Construction and Working Principle of Electrodynamometer Type WattmeterNow let us look at constructional details of electrodynamometer. It consists of following parts There are two types of coils present in the electrodynamometer. They are :
Moving Coil Moving coil moves the pointer with the help of spring control instrument. A limited amount of current flows through the moving coil so as to avoid heating. So in order to limit the current we have connect the high value resistor in series with the moving coil. The moving is air cored and is mounted on a pivoted spindle and can moves freely. In electrodynamometer type wattmeter, moving coil works as pressure coil. Hence moving coil is connected across the voltage and thus the current flowing through this coil is always proportional to the voltage. Fixed Coil The fixed coil is divided into two equal parts and these are connected in series with the load, therefore the load current will flow through these coils. Now the reason is very obvious of using two fixed coils instead of one, so that it can be constructed to carry considerable amount of electric current. These coils are called the current coils of electrodynamometer type wattmeter. Earlier these fixed coils are designed to carry the current of about 100 amperes but now the modern wattmeter are designed to carry current of about 20 amperes in order to save power. Control System Out of two controlling systems i.e.
- Gravity control
- Spring control, only spring controlled systems are used in these types of wattmeter. Gravity controlled system cannot be employed because they will appreciable amount of errors.
Advantages of Electrodynamometer Type WattmeterFollowing are the advantages of electrodynamometer type wattmeters and they are written as follows:
- Scale is uniform upto certain limit.
- They can be used for both to measure ac as well dc quantities as scale is calibrated for both.
Errors in Electrodynamometer Type WattmeterFollowing are the errors in the electrodynamometer type wattmeters:
- Errors in the pressure coil inductance.
- Errors may be due to pressure coil capacitance.
- Errors may be due to mutual inductance effects.
- Errors may be due connections.(i.e. pressure coil is connected after current coil)
- Error due to Eddy currents.
- Errors caused by vibration of moving system.
- Temperature error.
- Errors due to stray magnetic field.
Closely Related Articles Digital Frequency MeterElectrical Measuring Instruments | Types Accuracy Precision Resolution SpeedTransducer | Types of TransducerCathode Ray Oscilloscope | CROOhmmeter Working Principle of OhmmeterErrors in Measurement | Classification of ErrorsLissajous Patterns of CRO or Cathode Ray OscilloscopeFrequency Limitation of an OscilloscopePotentiometer Working Principle of PotentiometerPermanent Magnet Moving Coil Instrument or PMMC InstrumentCharacteristics of SensorsSignal GeneratorMoving Iron InstrumentInduction Type MetersMeasurement of Voltage Current and Frequency by OscilloscopeElectrostatic Type Instruments Construction Principle Torque EquationWhat is Low Power Factor Wattmeter ?Resistance Temperature Detector or RTD | Construction and Working PrincipleEnergy Meter with Lag Adjustment DevicesRectifier Type Instrument | Construction Principle of OperationThermistor Thermometer | Thermistor Temperature Sensor | Construction and Principle Digital Storage OscilloscopeThermocouple type Instruments Construction Principle of OperationThermistor Definition Properties Construction Characteristics and Applications Double Beam OscilloscopeProtection of MetersMeasurement of Three Phase PowerBimetallic Strip ThermometerSampling OscilloscopeVarmeter | Single Phase and Polyphase VarmeterThermocouple Temperature MeasurementTemperature TransducersMegger | Working Principle Types History Uses of MeggerRadiation Pyrometer | Types Working PrincipleOptical Pyrometer | Construction and Working PrinciplePiezoelectric TransducerStrain GaugeWeston Type Frequency MeterPower Factor Meters | Electrodynamometer Type Power Factor MeterInductive TransducersPhase Sequence IndicatorAmmeter Working Principle and Types of AmmeterLinear Variable Differential Transformer LVDTOscillator TransducerDigital MultimeterInsulation Resistance Test and Polarization Index TestHartley OscillatorTan Delta Test |Loss Angle Test | Dissipation Factor TestColpitts OscillatorElectronic DC VoltmeterWorking Principle of Voltmeter and Types of VoltmeterClapp OscillatorRC Phase Shift OscillatorDigital Voltmeters Working Principle of Digital VoltmeterWien Bridge OscillatorMeasurement of Electrical EnergyGunn OscillatorEnergy Meter TestingCrystal OscillatorAC PotentiometerMonostable MultivibratorConstruction of AC Energy MeterBistable MultivibratorWatt Hour MeterCapacitance MeterWhat is an Oscillator?Vector Impedance MeterAstable MultivibratorHow to Use a Digital Multimeter?Voltage Controlled Oscillator | VCOMore Related Articles Digital PotentiometersWheatstone Bridge Circuit Theory and PrincipleKelvin Bridge Circuit | Kelvin Double BridgeMaxwell Bridge Inductance Capacitance BridgeAnderson′s Bridge | Advantages Disadvantages of Anderson′s BridgeHay′s Bridge Circuit Theory Phasor Diagram Advantages ApplicationsOwens Bridge Circuit and AdvantagesSchering Bridge Measurement of Capacitance using Schering BridgeDe Sauty BridgeHeaviside Bridge CircuitBlavier Test | Murray Loop Test | Varley Loop Test | Fisher Loop TestSensor | Types of SensorTemperature Sensor Temperature Measurement | Types of Temperature SensorVoltage SensorMeasurement of ResistanceAir MeterWater MeterFlow MeasurementFlow MeterNew Articles Acidity Test of Transformer Insulating OilMagnetic FluxRing CounterDischarging a CapacitorCharging a Capacitor