# Electrodynamometer Type Wattmeter

**electrodynamometer wattmeter**, it very essential to know the principle of working of electrodynamometer type wattmeter. Dynamometer type wattmeter works on very simple principle and this principle can be stated as when any current carrying conductor is placed inside a magnetic field, it experiences a mechanical force and due this mechanical force deflection of conductor takes place.

## Construction and Working Principle of Electrodynamometer Type Wattmeter

Now let us look at constructional details of electrodynamometer. It consists of following partsThere are two types of coils present in the electrodynamometer. They are :

Moving Coil

Moving coil moves the pointer with the help of spring control instrument. A limited amount of current flows through the moving coil so as to avoid heating. So in order to limit the current we have connect the high value resistor in series with the moving coil. The moving is air cored and is mounted on a pivoted spindle and can moves freely. In **electrodynamometer type wattmeter**, moving coil works as pressure coil. Hence moving coil is connected across the voltage and thus the current flowing through this coil is always proportional to the voltage.

Fixed Coil

The fixed coil is divided into two equal parts and these are connected in series with the load, therefore the load current will flow through these coils. Now the reason is very obvious of using two fixed coils instead of one, so that it can be constructed to carry considerable amount of electric current. These coils are called the current coils of **electrodynamometer type wattmeter**. Earlier these fixed coils are designed to carry the current of about 100 amperes but now the modern wattmeter are designed to carry current of about 20 amperes in order to save power.

Control System

Out of two controlling systems i.e.

- Gravity control
- Spring control, only spring controlled systems are used in these types of wattmeter. Gravity controlled system cannot be employed because they will appreciable amount of errors.

Air friction damping is used, as eddy current damping will distort the weak operating magnetic field and thus it may leads to error.

Scale

There is uniform scale is used in these types of instrument as moving coil moves linearly over a range of 40 degrees to 50 degrees on either sides.

Now let us derive the expressions for the controlling torque and deflecting torques. In order to derive these expressions let us consider the circuit diagram given below: We know that instantaneous torque in electrodynamic type instruments is directly proportional to product of instantaneous values of currents flowing through both the coils and the rate of change of flux linked with the circuit.

Let I

_{1}and I

_{2}be the instantaneous values of currents in pressure and current coils respectively. So the expression for the torque can be written as: Where, x is the angle.

Now let the applied value of voltage across the pressure coil be Assuming the electrical resistance of the pressure coil be very high hence we can neglect reactance with respect to its resistance. In this the impedance is equal to its electrical resistance therefore it is purely resistive.

The expression for instantaneous current can be written as I

_{2}= v / R

_{p}where R

_{p}is the resistance of pressure coil. If there is phase difference between voltage and electric current, then expression for instantaneous current through current coil can be written as As current through the pressure coil in very very small compare to current through current coil hence current through the current coil can be considered as equal to total load current.

Hence the instantaneous value of torque can be written as Average value of deflecting torque can be obtained by integrating the instantaneous torque from limit 0 to T, where T is the time period of the cycle. Controlling torque is given by T

_{c}= K

_{x}where K is spring constant and x is final steady state value of deflection.

### Advantages of Electrodynamometer Type Wattmeter

Following are the**advantages of electrodynamometer type wattmeters**and they are written as follows:

- Scale is uniform upto certain limit.
- They can be used for both to measure ac as well dc quantities as scale is calibrated for both.

### Errors in Electrodynamometer Type Wattmeter

Following are the errors in the electrodynamometer type wattmeters:- Errors in the pressure coil inductance.
- Errors may be due to pressure coil capacitance.
- Errors may be due to mutual inductance effects.
- Errors may be due connections.(i.e. pressure coil is connected after current coil)
- Error due to Eddy currents.
- Errors caused by vibration of moving system.
- Temperature error.
- Errors due to stray magnetic field.

**Comments/Feedbacks**

Closely Related Articles Digital Frequency MeterElectrical Measuring Instruments | Types Accuracy Precision Resolution SpeedTransducer | Types of TransducerCathode Ray Oscilloscope | CROOhmmeter Working Principle of OhmmeterErrors in Measurement | Classification of ErrorsLissajous Patterns of CRO or Cathode Ray OscilloscopeFrequency Limitation of an OscilloscopePotentiometer Working Principle of PotentiometerPermanent Magnet Moving Coil Instrument or PMMC InstrumentCharacteristics of SensorsSignal GeneratorMoving Iron InstrumentInduction Type MetersMeasurement of Voltage Current and Frequency by OscilloscopeElectrostatic Type Instruments Construction Principle Torque EquationWhat is Low Power Factor Wattmeter ?Resistance Temperature Detector or RTD | Construction and Working PrincipleEnergy Meter with Lag Adjustment DevicesRectifier Type Instrument | Construction Principle of OperationThermistor Thermometer | Thermistor Temperature Sensor | Construction and Principle Digital Storage OscilloscopeThermocouple type Instruments Construction Principle of OperationThermistor Definition Properties Construction Characteristics and Applications Double Beam OscilloscopeProtection of MetersMeasurement of Three Phase PowerBimetallic Strip ThermometerSampling OscilloscopeVarmeter | Single Phase and Polyphase VarmeterThermocouple Temperature MeasurementTemperature TransducersMegger | Working Principle Types History Uses of MeggerRadiation Pyrometer | Types Working PrincipleOptical Pyrometer | Construction and Working PrinciplePiezoelectric TransducerStrain GaugeWeston Type Frequency MeterPower Factor Meters | Electrodynamometer Type Power Factor MeterInductive TransducersPhase Sequence IndicatorAmmeter Working Principle and Types of AmmeterLinear Variable Differential Transformer LVDTOscillator TransducerDigital MultimeterInsulation Resistance Test and Polarization Index TestHartley OscillatorTan Delta Test |Loss Angle Test | Dissipation Factor TestColpitts OscillatorElectronic DC VoltmeterWorking Principle of Voltmeter and Types of VoltmeterClapp OscillatorRC Phase Shift OscillatorDigital Voltmeters Working Principle of Digital VoltmeterWien Bridge OscillatorMeasurement of Electrical EnergyGunn OscillatorEnergy Meter TestingCrystal OscillatorAC PotentiometerMonostable MultivibratorConstruction of AC Energy MeterBistable MultivibratorWatt Hour MeterCapacitance MeterWhat is an Oscillator?Vector Impedance MeterAstable MultivibratorHow to Use a Digital Multimeter?Voltage Controlled Oscillator | VCOMore Related Articles Digital PotentiometersWheatstone Bridge Circuit Theory and PrincipleKelvin Bridge Circuit | Kelvin Double BridgeMaxwell Bridge Inductance Capacitance BridgeAnderson′s Bridge | Advantages Disadvantages of Anderson′s BridgeHay′s Bridge Circuit Theory Phasor Diagram Advantages ApplicationsOwens Bridge Circuit and AdvantagesSchering Bridge Measurement of Capacitance using Schering BridgeDe Sauty BridgeHeaviside Bridge CircuitBlavier Test | Murray Loop Test | Varley Loop Test | Fisher Loop TestSensor | Types of SensorTemperature Sensor Temperature Measurement | Types of Temperature SensorVoltage SensorMeasurement of ResistanceAir MeterWater MeterFlow MeasurementFlow MeterNew Articles Acidity Test of Transformer Insulating OilMagnetic FluxRing CounterDischarging a CapacitorCharging a Capacitor