# Characteristics of Sensors

Sensors are characterized depending on the value of some of the parameters. Important

**characteristics of sensors**and transducers are listed below:- Input characteristics
- Transfer characteristics
- Output characteristics

## Input Characteristics of Sensors

- Range: It is the minimum and maximum value of physical variable that the sensor can sense or measure. For example, a Resistance Temperature Detector (RTD) for the measurement of temperature has a range of -200 to 800
^{o}C. - Span: It is the difference between the maximum and minimum values of input. In above example, the span of RTD is 800 - (-200) = 1000
^{o}C. - Accuracy: The error in measurement is specified in terms of accuracy. It is defined as the difference between measured value and true value. It is defined in terms of % of full scale or % of reading.
X
_{t}is calculated by taking mean of infinite number of measurements. - Precision: It is defined as the closeness among a set of values. It is different from accuracy. Let X
_{t}be the true value of the variable X and a random experiment measures X_{1}, X_{2}, …. X_{i}as the value of X. We will say our measurements X_{1}, X_{2},… X_{i}are precise when they are very near to each other but not necessarily close to true value X_{t}. However, if we say X_{1}, X_{2},… X_{i}are accurate, it means that they are close to true value X_{t}and hence they are also close to each other. Hence accurate measurements are always precise. - Sensitivity: It is the ratio of change in output to change in input. If Y be the output quantity in response to input X, then sensitivity S can be expressed as
- Linearity: Linearity is the maximum deviation between the measured values of a sensor from ideal curve.
- Hysteresis: It is the difference in output when input is varied in two ways- increasing and decreasing.
- Resolution: It is the minimum change in input that can be sensed by the sensor.
- Reproducibility: It is defined as the ability of sensor to produce the same output when same input is applied.
- Repeatability: It is defined as the ability of sensor to produce the same output every time when the same input is applied and all the physical and measurement conditions kept the same including the operator, instrument, ambient conditions etc.
- Response time: It is generally expressed as the time at which the output reaches a certain percentage (for instance, 95 %) of its final value, in response to a step change of the input.

**Comments/Feedbacks**

Closely Related Articles Digital Frequency MeterElectrical Measuring Instruments | Types Accuracy Precision Resolution SpeedTransducer | Types of TransducerCathode Ray Oscilloscope | CROOhmmeter Working Principle of OhmmeterErrors in Measurement | Classification of ErrorsLissajous Patterns of CRO or Cathode Ray OscilloscopeFrequency Limitation of an OscilloscopePotentiometer Working Principle of PotentiometerPermanent Magnet Moving Coil Instrument or PMMC InstrumentSignal GeneratorMoving Iron InstrumentInduction Type MetersMeasurement of Voltage Current and Frequency by OscilloscopeElectrostatic Type Instruments Construction Principle Torque EquationWhat is Low Power Factor Wattmeter ?Resistance Temperature Detector or RTD | Construction and Working PrincipleEnergy Meter with Lag Adjustment DevicesRectifier Type Instrument | Construction Principle of OperationThermistor Thermometer | Thermistor Temperature Sensor | Construction and Principle Digital Storage OscilloscopeElectrodynamometer Type WattmeterThermocouple type Instruments Construction Principle of OperationThermistor Definition Properties Construction Characteristics and Applications Double Beam OscilloscopeProtection of MetersMeasurement of Three Phase PowerBimetallic Strip ThermometerSampling OscilloscopeVarmeter | Single Phase and Polyphase VarmeterThermocouple Temperature MeasurementTemperature TransducersMegger | Working Principle Types History Uses of MeggerRadiation Pyrometer | Types Working PrincipleOptical Pyrometer | Construction and Working PrinciplePiezoelectric TransducerStrain GaugeWeston Type Frequency MeterPower Factor Meters | Electrodynamometer Type Power Factor MeterInductive TransducersPhase Sequence IndicatorAmmeter Working Principle and Types of AmmeterLinear Variable Differential Transformer LVDTOscillator TransducerDigital MultimeterInsulation Resistance Test and Polarization Index TestHartley OscillatorTan Delta Test |Loss Angle Test | Dissipation Factor TestColpitts OscillatorElectronic DC VoltmeterWorking Principle of Voltmeter and Types of VoltmeterClapp OscillatorRC Phase Shift OscillatorDigital Voltmeters Working Principle of Digital VoltmeterWien Bridge OscillatorMeasurement of Electrical EnergyGunn OscillatorEnergy Meter TestingCrystal OscillatorAC PotentiometerMonostable MultivibratorConstruction of AC Energy MeterBistable MultivibratorWatt Hour MeterCapacitance MeterWhat is an Oscillator?Vector Impedance MeterAstable MultivibratorHow to Use a Digital Multimeter?Voltage Controlled Oscillator | VCOMore Related Articles Digital PotentiometersWheatstone Bridge Circuit Theory and PrincipleKelvin Bridge Circuit | Kelvin Double BridgeMaxwell Bridge Inductance Capacitance BridgeAnderson′s Bridge | Advantages Disadvantages of Anderson′s BridgeHay′s Bridge Circuit Theory Phasor Diagram Advantages ApplicationsOwens Bridge Circuit and AdvantagesSchering Bridge Measurement of Capacitance using Schering BridgeDe Sauty BridgeHeaviside Bridge CircuitBlavier Test | Murray Loop Test | Varley Loop Test | Fisher Loop TestSensor | Types of SensorTemperature Sensor Temperature Measurement | Types of Temperature SensorVoltage SensorMeasurement of ResistanceAir MeterWater MeterFlow MeasurementFlow MeterNew Articles Principle of Water Content Test of Insulating OilCollecting Oil Sample from Oil Immersed Electrical EquipmentCauses of Insulating Oil DeteriorationAcidity Test of Transformer Insulating OilMagnetic Flux