# Vector Impedance Meter

**Vector Impedance Meter**is employed for measuring both the amplitude and phase angle of impedance (Z). Normally, in other measuring techniques of impedance, the individual values of resistance and reactance are obtained in rectangular form. That is . But here, the impedance can be obtained in polar form. That is |Z| and phase angle (θ) of impedance can be acquired by this meter. The circuit is shown below. Two resistors with equal resistance value are incorporated here. The voltage drop across R

_{AB}is E

_{AB}and that of R

_{BC}is E

_{BC}. Both the values are the same and it is equal to half the value of input voltage (E

_{AC}).

A variable standard resistance (R_{ST}) is connected in series with the impedance (Z_{X}) which the value has to be obtained. The equal deflection method is used for the determination of the magnitude of the unknown impedance. This is by achieving the equal voltage drops across the variable resistor and the impedance (E_{AD} = E_{CD}) and evaluating the calibrated standard resistor (here it is R_{ST}) which is also necessary for achieving this condition.
The phase angle of the impedance (θ) can be acquired from taking the voltage reading across BD. Here it is E_{BD}. The meter deflection will vary in accordance with the Q factor (quality factor) of the connected unknown impedance. The Vacuum Tube Voltmeter (VTVM) normally reads AC voltage which varies from 0V to maximum value. When the voltage reading is zero, the value of Q will be zero and the phase angle will be 0^{o}. When the voltage reading becomes maximum value, the value of Q will be infinite and the phase angle will be 90^{o}. The angle between E_{AB} and E_{AD} will be equal to θ/2 (half of the phase angle of the unknown impedance). This is because E_{AD} = E_{DC}.
We know that the voltage across A and B (E_{AB}) will be equal to half of the voltage across A and C (E_{AC} which is the input voltage). The reading of voltmeter, E_{DB} can be thus obtained in terms of θ/2. Hence, θ (phase angle) can be determined. The vector diagram is shown below.
For obtaining the first approximation of the magnitude and phase angle of the impedance, this method is preferred. For achieving more accuracy in measurement the commercial **vector impedance meter** is preferred.

## Commercial Vector Impedance Meter

Impedance can be straight away measured by using commercial vector impedance meter in polar form. Only a single balancing control is used in this for getting both the phase angle and magnitude of the impedance. This method can be used to determine any combination of resistance (R), Capacitance (C) and Inductance (L). In addition to this, it can measure complex impedances rather than pure elements (C, L or R). The main disadvantage in conventional bridge circuits like too much consecutive adjustments are eliminated here. The range of measurements of impedance is 0.5 to 100,000Ω over the range of frequency 30 Hz to 40 kHz when external oscillator is used for giving the supply. The frequencies generated internally are of 1 kHz or 400 Hz or 60 Hz and externally up to 20 kHz. The accuracy in readings of the magnitude of impedance is ± 1% and for the phase angle it will be ± 2%. The circuit for the measurement of magnitude of impedance is shown below. Here, for the magnitude measurement R_{X}is the variable resistor and it can be altered with calibrating impedance dial. The voltage drops of both variable resistor and the unknown impedance (Z

_{X}) are made equal by adjusting this dial. Every voltage drop is made amplified by using the two modules of balanced amplifiers. This is then given to the section of the connected dual rectifier. In this, the arithmetical sum of the outputs of the rectifier can be obtained as zero and this is shown as the null reading in the indicating meter. Thus, the unknown impedance can be obtained directly from the dial of the variable resistor.

Next, we can see how the phase angle is obtained in this meter. First, the switch is set in the calibration position and the voltage injected is calibrated. This is done by setting it for getting the full scale deflection in the VTVM or indicating meter. After that, the function switch is kept in phase position. In this condition, the function switch will make the output of the balanced amplifier parallel before going to rectification. Now, the sum total of the AC voltages which is from the amplifiers is definitely a function of the vector difference among the AC voltages on the amplifiers. The voltage that is rectified as a result of this vector difference is indicated in the indicating meter or DC VTVM. This is actually the measure of the phase angle between the voltage drop across the unknown impedance and variable resistor. These voltage drops will be same in magnitude but phase is different. Hence, the phase angle is obtained by direct reading from this instrument. The quality factor and dissipation factor can also be calculated from this phase angle if needed. The circuit diagram for the measurement of phase angle (θ) is shown below.

Closely Related Articles Digital Frequency MeterOhmmeter Working Principle of OhmmeterInduction Type MetersWhat is Low Power Factor Wattmeter ?Electrodynamometer Type WattmeterProtection of MetersBimetallic Strip ThermometerVarmeter | Single Phase and Polyphase VarmeterWeston Type Frequency MeterAmmeter Working Principle and Types of AmmeterDigital MultimeterElectronic DC VoltmeterWorking Principle of Voltmeter and Types of VoltmeterDigital Voltmeters Working Principle of Digital VoltmeterConstruction of AC Energy MeterWatt Hour MeterCapacitance MeterHow to Use a Digital Multimeter?Flow MeasurementFlow MeterAir MeterWater MeterMore Related Articles Wheatstone Bridge Circuit Theory and PrincipleKelvin Bridge Circuit | Kelvin Double BridgeSensor | Types of SensorMaxwell Bridge Inductance Capacitance BridgeTemperature Sensor Temperature Measurement | Types of Temperature SensorAnderson′s Bridge | Advantages Disadvantages of Anderson′s BridgeHay′s Bridge Circuit Theory Phasor Diagram Advantages ApplicationsOwens Bridge Circuit and AdvantagesSchering Bridge Measurement of Capacitance using Schering BridgeDe Sauty BridgeHeaviside Bridge CircuitBlavier Test | Murray Loop Test | Varley Loop Test | Fisher Loop TestCathode Ray Oscilloscope | CROLissajous Patterns of CRO or Cathode Ray OscilloscopeMeasurement of Voltage Current and Frequency by OscilloscopeDigital Storage OscilloscopeDouble Beam OscilloscopeSampling OscilloscopeMeasurement of ResistanceFrequency Limitation of an OscilloscopeSignal GeneratorHartley OscillatorColpitts OscillatorClapp OscillatorRC Phase Shift OscillatorWien Bridge OscillatorGunn OscillatorCrystal OscillatorWhat is an Oscillator?Voltage Controlled Oscillator | VCOPotentiometer Working Principle of PotentiometerAC PotentiometerDigital PotentiometersElectrical Measuring Instruments | Types Accuracy Precision Resolution SpeedTransducer | Types of TransducerErrors in Measurement | Classification of ErrorsPermanent Magnet Moving Coil Instrument or PMMC InstrumentCharacteristics of SensorsMoving Iron InstrumentElectrostatic Type Instruments Construction Principle Torque EquationResistance Temperature Detector or RTD | Construction and Working PrincipleEnergy Meter with Lag Adjustment DevicesRectifier Type Instrument | Construction Principle of OperationThermistor Thermometer | Thermistor Temperature Sensor | Construction and Principle Thermocouple Type Instruments Construction Principle of OperationThermistor Definition Properties Construction Characteristics and Applications Measurement of Three Phase PowerThermocouple Temperature MeasurementTemperature TransducersMegger | Working Principle Types History Uses of MeggerRadiation Pyrometer | Types Working PrincipleOptical Pyrometer | Construction and Working PrinciplePiezoelectric TransducerStrain GaugePower Factor Meters | Electrodynamometer Type Power Factor MeterInductive TransducersPhase Sequence IndicatorLinear Variable Differential Transformer LVDTOscillator TransducerPolarization Index Test or PI TestVoltage SensorTan Delta Test |Loss Angle Test | Dissipation Factor TestMeasurement of Electrical EnergyEnergy Meter TestingMonostable MultivibratorBistable MultivibratorAstable MultivibratorNew Articles Series and Parallel Inductors Electric PowerMeasurement of Losses in Shunt ReactorThree Phase Shunt ReactorMeasurement of Insulation ResistanceAmpere's Circuital Law