Optical Pyrometer | Construction and Working Principleon 24/2/2012 & Updated on 29/8/2018
Construction of Optical PyrometerIt is quite simple. Consider it as a cylinder, which has a lens in one end and in the other end there is an eye piece. In between there is a lamp. In front of the eye piece there is a colored glass (usually red), to make lights monochromatic. The lamp is connected to a battery source through an ammeter and a rheostat as shown in the figure.
The optical pyrometer works in a certain simple process. The process is, the brightness of the filament of the lamp, that we are using through a battery source can be controlled by the rheostat. Now by controlling the incoming current, the brightness of the filament is increased or decreased. Going through this process there will be a certain point, when the filament of the lamp will not be visible from the eye piece. That very moment the brightness of the filament matches with the brightness of the hot body as seen through the monochromatic glass. From the reading of the ammeter of that particular condition we can get the temperature of the hot body, as the ammeter is previously calibrated in temperature scale.
Limitations of Optical PyrometerThere are some limitations of this pyrometer. Such as:--
- This type of pyrometer can measure the temperature of only those objects which are emitting light that means glowing objects.
- The optical pyrometer has a range of measuring temperature of 1400oC to near about 3500oC.
Rating = 5 & Total votes = 1