Voltage in Parallel
Voltage in Series
Voltage Regulator 7805
How to Use a Digital Multimeter?Closely Related Articles
Digital Frequency Meter
Ohmmeter Working Principle of Ohmmeter
Potentiometer Working Principle of Potentiometer
Induction Type Meters
What is Low Power Factor Wattmeter ?
Energy Meter with Lag Adjustment Devices
Electrodynamometer Type Wattmeter
Measurement of Three Phase Power
Varmeter | Single Phase and Polyphase Varmeter
Megger | Working Principle Types History Uses of Megger
Weston Type Frequency Meter
Power Factor Meters | Electrodynamometer Type Power Factor Meter
Phase Sequence Indicator
Ammeter Working Principle and Types of Ammeter
Insulation Resistance Test and Polarization Index Test
Tan Delta Test |Loss Angle Test | Dissipation Factor Test
Electronic DC Voltmeter
Working Principle of Voltmeter and Types of Voltmeter
Digital Voltmeters Working Principle of Digital Voltmeter
Measurement of Electrical Energy
Energy Meter Testing
Construction of AC Energy Meter
Watt Hour Meter
Vector Impedance Meter
How to Use a Digital Multimeter?
Electrical Measuring Instruments | Types Accuracy Precision Resolution Speed
Errors in Measurement | Classification of Errors
Permanent Magnet Moving Coil Instrument or PMMC Instrument
Moving Iron Instrument
Electrostatic Type Instruments Construction Principle Torque Equation
Rectifier Type Instrument | Construction Principle of Operation
Thermocouple type Instruments Construction Principle of Operation
Protection of Meters
Wheatstone Bridge Circuit Theory and Principle
Kelvin Bridge Circuit | Kelvin Double Bridge
Maxwell Bridge Inductance Capacitance Bridge
Anderson′s Bridge | Advantages Disadvantages of Anderson′s Bridge
Hay′s Bridge Circuit Theory Phasor Diagram Advantages Applications
Owens Bridge Circuit and Advantages
Schering Bridge Measurement of Capacitance using Schering Bridge
De Sauty Bridge
Heaviside Bridge Circuit
Blavier Test | Murray Loop Test | Varley Loop Test | Fisher Loop Test
Transducer | Types of Transducer
Radiation Pyrometer | Types Working Principle
Resistance Temperature Detector or RTD | Construction and Working Principle
Temperature Sensor Temperature Measurement | Types of Temperature Sensor
Sensor | Types of Sensor
Optical Pyrometer | Construction and Working Principle
Bimetallic Strip Thermometer
Thermistor Thermometer | Thermistor Temperature Sensor | Construction and Principle
Thermocouple Temperature Measurement
Linear Variable Differential Transformer LVDT
Thermistor Definition Properties Construction Characteristics and Applications
RC Phase Shift Oscillator
Wien Bridge Oscillator
555 Timer and 555 Timer Working
What is an Oscillator?
Voltage Controlled Oscillator | VCO
Cathode Ray Oscilloscope | CRO
Lissajous Patterns of CRO or Cathode Ray Oscilloscope
Frequency Limitation of an Oscilloscope
Measurement of Voltage Current and Frequency by Oscilloscope
Digital Storage Oscilloscope
Double Beam Oscilloscope
How to Use a Digital Multimeter?
A typical digital multimeter will be as shown by Figure 1 and comprises of selection switch, display, ports and probes as its major parts. Here the probes are to be inserted into appropriate ports and are to be connected across the parameter which needs to be inspected. Meanwhile one should ensure that the selection switch is to be kept at the position which is appropriate for the measurement. When done so, the multimeter displays the value of the parameter which is being analyzed. Generally digital multimeters are used to measure three important parameters viz., current, voltage and resistance. Apart from these, they can also be used to perform special functions like diode check, capacitance measurement, Transistor hFE or DC current gain, frequency measurement and continuity check. In this article, we present a brief note on the most frequently used applications of multimeter which are for current, voltage and resistance measurement along with diode and continuity checks.
Current Measurement by Using a Digital MultimeterUnder this category, the digital multimeter mimics the behavior of an ammeter as it is used to measure current. To accomplish this, insert the red probe of the multimeter to one of the current measuring sockets: mA (to measure low level of current) or 20 A (to measure larger current). Connect the meter along the line through which the current is to be measured (nothing but series connection). Next set an approximate range around which we expect the current to be in the ammeter section of Figure 1. In this state, if we switch on the power supply, then the meter will read the current flowing through the circuit.
Voltage Measurement by Using a Digital MultimeterWhen set to measure the voltage, the multimeter acts like a voltmeter. To start with, one has to insert the red and the black probes of the multimeter to the sockets marked as ‘V’ and ‘COM’, respectively. Then we have to select the expected range in which our voltage would be. Simultaneously, even AC or DC should also be selected in the voltmeter section of Figure 1. On doing so, the meter reads the value of the voltage, provided one connects the leads across the component (in parallel fashion) or at the point at which the voltage needs to be measured.
Resistance Measurement by Using a Digital MultimeterIn this case, we configure the multimeter to act like an ohmmeter. Here the red and the black probes of the multimeter are inserted into the sockets marked as ‘V’ and ‘COM’, respectively while the selection switch is set to an expected range in ohmmeter region (Figure 1). Now, the leads need to be connected across the component whose resistance is to be known. On doing so, we get a reading in the display part of the multimeter which reads the value of the resistance.
Diode Check by Using a Digital MultimeterFor this case, insert the probes into the sockets as that in the case of voltage measurement and set the selection switch to point towards diode check position shown in Figure 1. Now when the red lead of the multimeter is connected to positive terminal of the diode while its negative lead is connected to the negative terminal of the diode, then we have to get a low reading on the multimeter. On the other hand, if we connect the red lead to the negative terminal of the diode and the black to the positive terminal, then we have to get a high value. If the readings obtained are as per our expectation, then we say that the diode is working properly; else no. More information regarding this can be found in the article “Diode testing”.
Continuity Check by Using a Digital MultimeterContinuity check is used to know whether there exists any low resistance path via two points i.e. to check whether the points are short or not. To accomplish this task, the probes are inserted into the sockets as that in the case of voltage measurement and selector switch is made to point towards continuity check position (Figure 1). Then, the points to be tested are touched with the leads of the probes. Now, if the multimeter beeps out, then it means that the points are shorted or else the resistance between them can be read out from the display.
Closely Related Articles Digital Frequency MeterOhmmeter Working Principle of OhmmeterPotentiometer Working Principle of PotentiometerInduction Type MetersWhat is Low Power Factor Wattmeter ?Energy Meter with Lag Adjustment DevicesElectrodynamometer Type WattmeterMeasurement of Three Phase PowerVarmeter | Single Phase and Polyphase VarmeterMegger | Working Principle Types History Uses of MeggerWeston Type Frequency MeterPower Factor Meters | Electrodynamometer Type Power Factor MeterPhase Sequence IndicatorAmmeter Working Principle and Types of AmmeterDigital MultimeterInsulation Resistance Test and Polarization Index TestTan Delta Test |Loss Angle Test | Dissipation Factor TestElectronic DC VoltmeterWorking Principle of Voltmeter and Types of VoltmeterDigital Voltmeters Working Principle of Digital VoltmeterMeasurement of Electrical EnergyEnergy Meter TestingAC PotentiometerConstruction of AC Energy MeterWatt Hour MeterCapacitance MeterVector Impedance MeterFlow MeasurementMore Related Articles Electrical Measuring Instruments | Types Accuracy Precision Resolution SpeedErrors in Measurement | Classification of ErrorsPermanent Magnet Moving Coil Instrument or PMMC InstrumentMoving Iron InstrumentElectrostatic Type Instruments Construction Principle Torque EquationRectifier Type Instrument | Construction Principle of OperationThermocouple type Instruments Construction Principle of OperationProtection of MetersWheatstone Bridge Circuit Theory and PrincipleKelvin Bridge Circuit | Kelvin Double BridgeMaxwell Bridge Inductance Capacitance BridgeAnderson′s Bridge | Advantages Disadvantages of Anderson′s BridgeHay′s Bridge Circuit Theory Phasor Diagram Advantages ApplicationsOwens Bridge Circuit and AdvantagesSchering Bridge Measurement of Capacitance using Schering BridgeDe Sauty BridgeHeaviside Bridge CircuitBlavier Test | Murray Loop Test | Varley Loop Test | Fisher Loop TestTransducer | Types of TransducerRadiation Pyrometer | Types Working PrincipleResistance Temperature Detector or RTD | Construction and Working PrincipleStrain GaugeTemperature Sensor Temperature Measurement | Types of Temperature SensorSensor | Types of SensorOptical Pyrometer | Construction and Working PrincipleBimetallic Strip ThermometerPiezoelectric TransducerThermistor Thermometer | Thermistor Temperature Sensor | Construction and Principle Thermocouple Temperature MeasurementInductive TransducersTemperature TransducersLinear Variable Differential Transformer LVDTOscillator TransducerThermistor Definition Properties Construction Characteristics and Applications Hartley OscillatorColpitts OscillatorClapp OscillatorRC Phase Shift OscillatorWien Bridge OscillatorGunn OscillatorCrystal OscillatorMonostable MultivibratorBistable Multivibrator555 Timer and 555 Timer WorkingWhat is an Oscillator?Astable MultivibratorVoltage Controlled Oscillator | VCOCathode Ray Oscilloscope | CROLissajous Patterns of CRO or Cathode Ray OscilloscopeFrequency Limitation of an OscilloscopeSignal GeneratorMeasurement of Voltage Current and Frequency by OscilloscopeDigital Storage OscilloscopeDouble Beam OscilloscopeSampling OscilloscopeNew Articles Voltage SensorFlow MeasurementVoltage in ParallelVoltage in SeriesVoltage Regulator 7805How to Use a Digital Multimeter?