# Potentiometer Working Principle of Potentiometer

**working principle of potentiometer**is very very simple. Suppose we have connected two battery in head to head and tale to tale through a galvanometer. That means the positive terminals of both battery are connected together and negative terminals are also connected together through a galvanometer as shown in the figure below. Here in the figure it is clear that if the voltage of both battery cells is exactly equal, there will be no circulating current in the circuit and hence the galvanometer shows null deflection. The

**working principle of potentiometer**depends upon this phenomenon.

Now let's think about another circuit, where a battery is connected across a resistor via a switch and a rheostat as shown in the figure below, there will be a voltage drop across the resistor. As there is a voltage drop across the resistor, this portion of the circuit can be considered as a voltage source for other external circuits. That means anything connected across the resistor will get voltage. If the resistor has uniform cross section throughout its length, the electrical resistance per unit length of the resistor is also uniform throughout its length. Hence, voltage drop per unit length of the resistor is also uniform. Suppose the current through the resistor is i A and resistance per unit length of the resistor is r Ω. Then the voltage appears per unit length across the resistor would be 'ir' ans say it is v volt.

Now, positive terminal of a standard cell is connected to point A on the sliding resistor and negative terminal of the same is connected with a galvanometer. Other end of the galvanometer is in contact with the resistor via a sliding contact as shown in the figure above. By adjusting this sliding end, a point like B is found where, there is no current through the galvanometer, hence no deflection of galvanometer. That means emf of the standard cell is just balanced by the voltage drop appears across AB. Now if the distance between point A and B is L, then it can be written emf of standard cell E = Lv volt. As v (voltage drop per unit length of the sliding resistor) is known and L is measured from the scale attached to the resistor, the value of E i.e. emf of standard cell can also be calculated from the above simple equation very easily.We said earlier in this section that one of the uses of potentiometer is to compare emfs of different cells. Let's discuss how a DC potentiometer can compare emfs of two different cells. Let's think of two cells whose emf's are to be compared are joined as shown in the figure below. The positive terminals of the cells and source battery are joined together. The negative terminals of the cells are joined with the galvanometer in turn through a two way switch. The other end of the galvanometer is connected to a sliding contact on the resistor. Now by adjusting sliding contact on the resistor, it is found that the null deflection of galvanometer comes for first cell at a length of L on the scale and after positioning to way switch to second cell and then by adjusting the sliding contact, it is found that the null deflection of galvanometer comes for that cell at a length of L_{1} on the scale.Let's think of the first cell as standard cell and it's emf is E and second cell is unknown cell whose emf is E_{1}. Now as per above explanation,

E = Lv volt and

L_{1} = L_{1}v volt

Dividing one equation by other, we get

Closely Related Articles AC PotentiometerDigital PotentiometersMore Related Articles Wheatstone Bridge Circuit Theory and PrincipleKelvin Bridge Circuit | Kelvin Double BridgeSensor | Types of SensorMaxwell Bridge Inductance Capacitance BridgeTemperature Sensor Temperature Measurement | Types of Temperature SensorAnderson′s Bridge | Advantages Disadvantages of Anderson′s BridgeHay′s Bridge Circuit Theory Phasor Diagram Advantages ApplicationsOwens Bridge Circuit and AdvantagesSchering Bridge Measurement of Capacitance using Schering BridgeDe Sauty BridgeHeaviside Bridge CircuitBlavier Test | Murray Loop Test | Varley Loop Test | Fisher Loop TestCathode Ray Oscilloscope | CROLissajous Patterns of CRO or Cathode Ray OscilloscopeMeasurement of Voltage Current and Frequency by OscilloscopeDigital Storage OscilloscopeDouble Beam OscilloscopeSampling OscilloscopeMeasurement of ResistanceDigital Frequency MeterOhmmeter Working Principle of OhmmeterInduction Type MetersWhat is Low Power Factor Wattmeter ?Electrodynamometer Type WattmeterProtection of MetersBimetallic Strip ThermometerVarmeter | Single Phase and Polyphase VarmeterWeston Type Frequency MeterAmmeter Working Principle and Types of AmmeterDigital MultimeterElectronic DC VoltmeterWorking Principle of Voltmeter and Types of VoltmeterDigital Voltmeters Working Principle of Digital VoltmeterConstruction of AC Energy MeterWatt Hour MeterCapacitance MeterVector Impedance MeterHow to Use a Digital Multimeter?Flow MeasurementFlow MeterAir MeterWater MeterFrequency Limitation of an OscilloscopeSignal GeneratorHartley OscillatorColpitts OscillatorClapp OscillatorRC Phase Shift OscillatorWien Bridge OscillatorGunn OscillatorCrystal OscillatorWhat is an Oscillator?Voltage Controlled Oscillator | VCOElectrical Measuring Instruments | Types Accuracy Precision Resolution SpeedTransducer | Types of TransducerErrors in Measurement | Classification of ErrorsPermanent Magnet Moving Coil Instrument or PMMC InstrumentCharacteristics of SensorsMoving Iron InstrumentElectrostatic Type Instruments Construction Principle Torque EquationResistance Temperature Detector or RTD | Construction and Working PrincipleEnergy Meter with Lag Adjustment DevicesRectifier Type Instrument | Construction Principle of OperationThermistor Thermometer | Thermistor Temperature Sensor | Construction and Principle Thermocouple Type Instruments Construction Principle of OperationThermistor Definition Properties Construction Characteristics and Applications Measurement of Three Phase PowerThermocouple Temperature MeasurementTemperature TransducersMegger | Working Principle Types History Uses of MeggerRadiation Pyrometer | Types Working PrincipleOptical Pyrometer | Construction and Working PrinciplePiezoelectric TransducerStrain GaugePower Factor Meters | Electrodynamometer Type Power Factor MeterInductive TransducersPhase Sequence IndicatorLinear Variable Differential Transformer LVDTOscillator TransducerPolarization Index Test or PI TestVoltage SensorTan Delta Test |Loss Angle Test | Dissipation Factor TestMeasurement of Electrical EnergyEnergy Meter TestingMonostable MultivibratorBistable MultivibratorAstable MultivibratorNew Articles Series and Parallel Inductors Electric PowerMeasurement of Losses in Shunt ReactorThree Phase Shunt ReactorMeasurement of Insulation ResistanceAmpere's Circuital Law