# Potentiometer Working Principle of Potentiometer

**working principle of potentiometer**is very very simple. Suppose we have connected two battery in head to head and tale to tale through a galvanometer. That means the positive terminals of both battery are connected together and negative terminals are also connected together through a galvanometer as shown in the figure below. Here in the figure it is clear that if the voltage of both battery cells is exactly equal, there will be no circulating current in the circuit and hence the galvanometer shows null deflection. The

**working principle of potentiometer**depends upon this phenomenon.

Now let's think about another circuit, where a battery is connected across a resistor via a switch and a rheostat as shown in the figure below, there will be a voltage drop across the resistor. As there is a voltage drop across the resistor, this portion of the circuit can be considered as a voltage source for other external circuits. That means anything connected across the resistor will get voltage. If the resistor has uniform cross section throughout its length, the electrical resistance per unit length of the resistor is also uniform throughout its length. Hence, voltage drop per unit length of the resistor is also uniform. Suppose the current through the resistor is i A and resistance per unit length of the resistor is r Ω. Then the voltage appears per unit length across the resistor would be 'ir' ans say it is v volt.

Now, positive terminal of a standard cell is connected to point A on the sliding resistor and negative terminal of the same is connected with a galvanometer. Other end of the galvanometer is in contact with the resistor via a sliding contact as shown in the figure above. By adjusting this sliding end, a point like B is found where, there is no current through the galvanometer, hence no deflection of galvanometer. That means emf of the standard cell is just balanced by the voltage drop appears across AB. Now if the distance between point A and B is L, then it can be written emf of standard cell E = Lv volt. As v (voltage drop per unit length of the sliding resistor) is known and L is measured from the scale attached to the resistor, the value of E i.e. emf of standard cell can also be calculated from the above simple equation very easily.We said earlier in this section that one of the uses of potentiometer is to compare emfs of different cells. Let's discuss how a DC potentiometer can compare emfs of two different cells. Let's think of two cells whose emf's are to be compared are joined as shown in the figure below. The positive terminals of the cells and source battery are joined together. The negative terminals of the cells are joined with the galvanometer in turn through a two way switch. The other end of the galvanometer is connected to a sliding contact on the resistor. Now by adjusting sliding contact on the resistor, it is found that the null deflection of galvanometer comes for first cell at a length of L on the scale and after positioning to way switch to second cell and then by adjusting the sliding contact, it is found that the null deflection of galvanometer comes for that cell at a length of L_{1} on the scale.Let's think of the first cell as standard cell and it's emf is E and second cell is unknown cell whose emf is E_{1}. Now as per above explanation,

E = Lv volt and

L_{1} = L_{1}v volt

Dividing one equation by other, we get

Closely Related Articles Digital Frequency MeterOhmmeter Working Principle of OhmmeterInduction Type MetersWhat is Low Power Factor Wattmeter ?Energy Meter with Lag Adjustment DevicesElectrodynamometer Type WattmeterMeasurement of Three Phase PowerVarmeter | Single Phase and Polyphase VarmeterMegger | Working Principle Types History Uses of MeggerWeston Type Frequency MeterPower Factor Meters | Electrodynamometer Type Power Factor MeterPhase Sequence IndicatorAmmeter Working Principle and Types of AmmeterDigital MultimeterInsulation Resistance Test and Polarization Index TestTan Delta Test |Loss Angle Test | Dissipation Factor TestElectronic DC VoltmeterWorking Principle of Voltmeter and Types of VoltmeterDigital Voltmeters Working Principle of Digital VoltmeterMeasurement of Electrical EnergyEnergy Meter TestingAC PotentiometerConstruction of AC Energy MeterWatt Hour MeterCapacitance MeterVector Impedance MeterHow to Use a Digital Multimeter?More Related Articles Electrical Measuring Instruments | Types Accuracy Precision Resolution SpeedErrors in Measurement | Classification of ErrorsPermanent Magnet Moving Coil Instrument or PMMC InstrumentMoving Iron InstrumentElectrostatic Type Instruments Construction Principle Torque EquationRectifier Type Instrument | Construction Principle of OperationThermocouple type Instruments Construction Principle of OperationProtection of MetersWheatstone Bridge Circuit Theory and PrincipleKelvin Bridge Circuit | Kelvin Double BridgeMaxwell Bridge Inductance Capacitance BridgeAnderson′s Bridge | Advantages Disadvantages of Anderson′s BridgeHay′s Bridge Circuit Theory Phasor Diagram Advantages ApplicationsOwens Bridge Circuit and AdvantagesSchering Bridge Measurement of Capacitance using Schering BridgeDe Sauty BridgeHeaviside Bridge CircuitBlavier Test | Murray Loop Test | Varley Loop Test | Fisher Loop TestTransducer | Types of TransducerRadiation Pyrometer | Types Working PrincipleResistance Temperature Detector or RTD | Construction and Working PrincipleStrain GaugeTemperature Sensor Temperature Measurement | Types of Temperature SensorSensor | Types of SensorOptical Pyrometer | Construction and Working PrincipleBimetallic Strip ThermometerPiezoelectric TransducerThermistor Thermometer | Thermistor Temperature Sensor | Construction and Principle Thermocouple Temperature MeasurementInductive TransducersTemperature TransducersLinear Variable Differential Transformer LVDTOscillator TransducerThermistor Definition Properties Construction Characteristics and Applications Hartley OscillatorColpitts OscillatorClapp OscillatorRC Phase Shift OscillatorWien Bridge OscillatorGunn OscillatorCrystal OscillatorMonostable MultivibratorBistable Multivibrator555 Timer and 555 Timer WorkingWhat is an Oscillator?Astable MultivibratorVoltage Controlled Oscillator | VCOCathode Ray Oscilloscope | CROLissajous Patterns of CRO or Cathode Ray OscilloscopeFrequency Limitation of an OscilloscopeSignal GeneratorMeasurement of Voltage Current and Frequency by OscilloscopeDigital Storage OscilloscopeDouble Beam OscilloscopeSampling OscilloscopeNew Articles Voltage in ParallelVoltage in SeriesVoltage Regulator 7805How to Use a Digital Multimeter?Vector Impedance MeterVoltage Follower